title: On the precision of time-of-flight shear wave speed estimation in homogeneous soft solids: initial results using a matrix array transducer.
abstract: A system capable of tracking radiation-force-induced shear wave propagation in a 3-D volume using ultrasound is presented. In contrast to existing systems, which use 1-D array transducers, a 2-D matrix array is used for tracking shear wave displacements. A separate single-element transducer is used for radiation force excitation. This system allows shear wave propagation in all directions away from the push to be observed. It is shown that for a limit of 64 tracking beams, by placing the beams at the edges of the measurement region of interest (ROI) at multiple directions from the push, time-of- flight (TOF) shear wave speed (SWS) measurement uncertainty can theoretically be reduced by 40% compared with equally spacing the tracking beams within the ROI along a single plane, as is typical when using a 1-D array for tracking. This was verified by simulation, and a reduction of 30% was experimentally observed on a homogeneous phantom. Analytical expressions are presented for the relationship between TOF SWS measurement uncertainty and various shear wave imaging parameters. It is shown that TOF SWS uncertainty is inversely proportional to ROI size, and inversely proportional to the square root of the number of tracking locations for a given distribution of beam locations relative to the push. TOF SWS uncertainty is shown to increase with the square of the SWS, indicating that TOF SWS measurements are intrinsically less precise for stiffer materials.
authors.affiliation: Department of Biomedical Engineering, Duke University, Durham, NC, USA. mhw12@duke.edu
authors.forename: Michael
authors.initials: M
authors.lastname: Wang
authors.forename: Brett
authors.initials: B
authors.lastname: Byram
authors.forename: Mark
authors.initials: M
authors.lastname: Palmeri
authors.forename: Ned
authors.initials: N
authors.lastname: Rouze
authors.forename: Kathryn
authors.initials: K
authors.lastname: Nightingale
mesh_terms: D003198:Computer Simulation; D007091:Image Processing, Computer-Assisted / Q000379:methods*; D019047:Phantoms, Imaging; D059629:Signal-To-Noise Ratio; D014159:Transducers; D014463:Ultrasonography / Q000295:instrumentation* / Q000379:methods
references.citation: Ultrasound Med Biol. 2010 May;36(5):802-13
references.pmid: 20381950
references.citation: Ultrasound Med Biol. 2003 Dec;29(12):1715-23
references.pmid: 14698339
references.citation: IEEE Trans Ultrason Ferroelectr Freq Control. 2005 Nov;52(11):1931-42
references.pmid: 16422405
references.citation: IEEE Trans Ultrason Ferroelectr Freq Control. 2007 Feb;54(2):301-12
references.pmid: 17328327
references.citation: IEEE Trans Ultrason Ferroelectr Freq Control. 2004 Apr;51(4):396-409
references.pmid: 15139541
references.citation: IEEE Trans Ultrason Ferroelectr Freq Control. 2005 Oct;52(10):1699-712
references.pmid: 16382621
references.citation: J Gastroenterol. 2012 Apr;47(4):461-9
references.pmid: 22223175
references.citation: IEEE Trans Med Imaging. 2012 Mar;31(3):554-62
references.pmid: 22020673
references.citation: IEEE Trans Ultrason Ferroelectr Freq Control. 2007 Sep;54(9):1751-71
references.pmid: 17941382
references.citation: IEEE Trans Ultrason Ferroelectr Freq Control. 1999;46(5):1057-67
references.pmid: 18244299
references.citation: J Acoust Soc Am. 2009 Nov;126(5):2733-45
references.pmid: 19894849
references.citation: Ultrasound Med Biol. 2003 Dec;29(12):1705-13
references.pmid: 14698338
references.citation: Ultrasound Med Biol. 1998 Nov;24(9):1419-35
references.pmid: 10385964
references.citation: IEEE Trans Ultrason Ferroelectr Freq Control. 2006 Jul;53(7):1300-13
references.pmid: 16889337
references.citation: Clin Gastroenterol Hepatol. 2007 Oct;5(10):1207-1213.e2
references.pmid: 17916548
references.citation: Ultrasound Med Biol. 2010 Oct;36(10):1662-76
references.pmid: 20800942
references.citation: Ultrasound Med Biol. 2008 Apr;34(4):546-58
references.pmid: 18222031
references.citation: Radiology. 2009 Aug;252(2):595-604
references.pmid: 19703889
references.citation: J Acoust Soc Am. 2004 Jun;115(6):2781-5
references.pmid: 15237800
references.citation: IEEE Trans Med Imaging. 2009 Mar;28(3):313-22
references.pmid: 19244004
references.citation: IEEE Trans Ultrason Ferroelectr Freq Control. 2011 Jun;58(6):1122-6
references.pmid: 21693392
references.citation: Science. 1995 Sep 29;269(5232):1854-7
references.pmid: 7569924
references.citation: NMR Biomed. 2006 Apr;19(2):173-9
references.pmid: 16521091
references.citation: IEEE Trans Ultrason Ferroelectr Freq Control. 2003 Jun;50(6):631-41
references.pmid: 12839175
references.citation: Ultrasound Med Biol. 2008 Sep;34(9):1373-86
references.pmid: 18395961
references.citation: Ultrasound Med Biol. 2011 Nov;37(11):1884-92
references.pmid: 21924817