title: Palm top plasma focus device as a portable pulsed neutron source.
abstract: Development of a palm top plasma focus device generating (5.2 ± 0.8) × 10(4) neutrons∕pulse into 4π steradians with a pulse width of 15 ± 3 ns is reported for the first time. The weight of the system is less than 1.5 kg. The system comprises a compact capacitor bank, a triggered open air spark gap switch, and a sealed type miniature plasma focus tube. The setup is around 14 cm in diameter and 12.5 cm in length. The energy driver for the unit is a capacitor bank of four cylindrical commercially available electrolytic capacitors. Each capacitor is of 2 μF capacity, 4.5 cm in diameter, and 9.8 cm in length. The cost of each capacitor is less than US$ 10. The internal diameter and the effective length of the plasma focus unit are 2.9 cm and 5 cm, respectively. A DC to DC converter power supply powered by two rechargeable batteries charges the capacitor bank to the desired voltage and also provides a trigger pulse of -15 kV to the spark gap. The maximum energy of operation of the device is 100 J (8 μF, 5 kV, 59 kA) with deuterium gas filling pressure of 3 mbar. The neutrons have also been produced at energy as low as 36 J (3 kV) of operation. The neutron diagnostics are carried out with a bank of (3)He detectors and with a plastic scintillator detector. The device is portable, reusable, and can be operated for multiple shots with a single gas filling.
authors.affiliation: Applied Physics Division, Bhabha Atomic Research Centre, Mumbai 400 085, India.
authors.forename: R K
authors.initials: RK
authors.lastname: Rout
authors.forename: Ram
authors.initials: R
authors.lastname: Niranjan
authors.forename: P
authors.initials: P
authors.lastname: Mishra
authors.forename: R
authors.initials: R
authors.lastname: Srivastava
authors.forename: A M
authors.initials: AM
authors.lastname: Rawool
authors.forename: T C
authors.initials: TC
authors.lastname: Kaushik
authors.forename: Satish C
authors.initials: SC
authors.lastname: Gupta