Anatomical noise in contrast-enhanced digital mammography. Part II. Dual-energy imaging.
OBJECTIVE
Dual-energy (DE) contrast-enhanced digital mammography (CEDM) uses an iodinated contrast agent in combination with digital mammography (DM) to evaluate lesions on the basis of tumor angiogenesis. In DE imaging, low-energy (LE) and high-energy (HE) images are acquired after contrast administration and their logarithms are subtracted to cancel the appearance of normal breast tissue. Often there is incomplete signal cancellation in the subtracted images, creating a background "clutter" that can impair lesion detection. This is the second component of a two-part report on anatomical noise in CEDM. In Part I the authors characterized the anatomical noise for single-energy (SE) temporal subtraction CEDM by a power law, with model parameters α and β. In this work the authors quantify the anatomical noise in DE CEDM clinical images and compare this with the noise in SE CEDM. The influence on the anatomical noise of the presence of iodine in the breast, the timing of imaging postcontrast administration, and the x-ray energy used for acquisition are each evaluated.


METHODS
The power law parameters, α and β, were measured from unprocessed LE and HE images and from DE subtracted images to quantify the anatomical noise. A total of 98 DE CEDM cases acquired in a previous clinical pilot study were assessed. Conventional DM images from 75 of the women were evaluated for comparison with DE CEDM. The influence of the imaging technique on anatomical noise was determined from an analysis of differences between the power law parameters as measured in DM, LE, HE, and DE subtracted images for each subject.


RESULTS
In DE CEDM, weighted image subtraction lowers β to about 1.1 from 3.2 and 3.1 in LE and HE unprocessed images, respectively. The presence of iodine has a small but significant effect in LE images, reducing β by about 0.07 compared to DM, with α unchanged. Increasing the x-ray energy, from that typical in DM to a HE beam, significantly decreases α by about 2×10(-5) mm2, and lowers β by about 0.14 compared to LE images. A comparison of SE and DE CEDM at 4 min postcontrast shows equivalent power law parameters in unprocessed images, and lower α and β by about 3×10(-5) mm2 and 0.50, respectively, in DE versus SE subtracted images.


CONCLUSIONS
Image subtraction in both SE and DE CEDM reduces β by over a factor of 2, while maintaining α below that in DM. Given the equivalent α between SE and DE unprocessed CEDM images, and the smaller anatomical noise in the DE subtracted images, the DE approach may have an advantage over SE CEDM. It will be necessary to test this potential advantage in future lesion detectability experiments, which account for realistic lesion signals. The authors' results suggest that LE images could be used in place of DM images in CEDM exam interpretation.
Sunnybrook Research Institute, 2075 Bayview Avenue, Toronto, Ontario M4N 3M5, Canada. melissa.hill@sri.utoronto.ca
Melissa L
ML
Hill
James G
JG
Mainprize
Ann-Katherine
AK
Carton
Sylvie
S
Saab-Puong
Razvan
R
Iordache
Serge
S
Muller
Roberta A
RA
Jong
Clarisse
C
Dromain
Martin J
MJ
Yaffe
D000328:Adult; D000368:Aged; D000369:Aged, 80 and over; D003287:Contrast Media; D005260:Female; D006801:Humans; D008327:Mammography / Q000379:methods*; D008875:Middle Aged; D011856:Radiographic Image Enhancement / Q000379:methods*; D059629:Signal-To-Noise Ratio
D003287:Contrast Media