Rapid detection of urinary polyomavirus BK by heterodyne-based surface plasmon resonance biosensor.
In renal transplant patients, immunosuppressive therapy may result in the reactivation of polyomavirus BK (BKV), leading to polyomavirus-associated nephropathy (PVAN), which inevitably causes allograft failure. Since the treatment outcomes of PVAN remain unsatisfactory, early identification and continuous monitoring of BKV reactivation and reduction of immunosuppressants are essential to prevent PVAN development. The present study demonstrated that the developed dual-channel heterodyne-based surface plasmon resonance (SPR) biosensor is applicable for the rapid detection of urinary BKV. The use of a symmetrical reference channel integrated with the poly(ethylene glycol)-based low-fouling self-assembled monolayer to reduce the environmental variations and the nonspecific noise was proven to enhance the sensitivity in urinary BKV detection. Experimentally, the detection limit of the biosensor for BKV detection was estimated to be around 8500  copies/mL. In addition, urine samples from five renal transplant patients were tested to rapidly distinguish PVAN-positive and PVAN-negative renal transplant patients. By virtue of its simplicity, rapidity, and applicability, the SPR biosensor is a remarkable potential to be used for continuous clinical monitoring of BKV reactivation.
Chang Gung University, Department of Electronic Engineering, Taoyuan 33002, Taiwan.
Li-Chen
LC
Su
Ya-Chung
YC
Tian
Ying-Feng
YF
Chang
Chien
C
Chou
Chao-Sung
CS
Lai
D001739:BK Virus / Q000737:chemistry / Q000276:immunology / Q000302:isolation & purification*; D006801:Humans; D007674:Kidney Diseases / Q000175:diagnosis / Q000652:urine* / Q000821:virology*; D016030:Kidney Transplantation; D057230:Limit of Detection; D027601:Polyomavirus Infections / Q000175:diagnosis / Q000652:urine* / Q000821:virology*; D020349:Surface Plasmon Resonance / Q000379:methods*; D014412:Tumor Virus Infections / Q000175:diagnosis / Q000652:urine / Q000821:virology