Influence of the partial volume correction method on (18)F-fluorodeoxyglucose brain kinetic modelling from dynamic PET images reconstructed with resolution model based OSEM.
Kinetic parameters estimated from dynamic (18)F-fluorodeoxyglucose ((18)F-FDG) PET acquisitions have been used frequently to assess brain function in humans. Neglecting partial volume correction (PVC) for a dynamic series has been shown to produce significant bias in model estimates. Accurate PVC requires a space-variant model describing the reconstructed image spatial point spread function (PSF) that accounts for resolution limitations, including non-uniformities across the field of view due to the parallax effect. For ordered subsets expectation maximization (OSEM), image resolution convergence is local and influenced significantly by the number of iterations, the count density, and background-to-target ratio. As both count density and background-to-target values for a brain structure can change during a dynamic scan, the local image resolution may also concurrently vary. When PVC is applied post-reconstruction the kinetic parameter estimates may be biased when neglecting the frame-dependent resolution. We explored the influence of the PVC method and implementation on kinetic parameters estimated by fitting (18)F-FDG dynamic data acquired on a dedicated brain PET scanner and reconstructed with and without PSF modelling in the OSEM algorithm. The performance of several PVC algorithms was quantified with a phantom experiment, an anthropomorphic Monte Carlo simulation, and a patient scan. Using the last frame reconstructed image only for regional spread function (RSF) generation, as opposed to computing RSFs for each frame independently, and applying perturbation geometric transfer matrix PVC with PSF based OSEM produced the lowest magnitude bias kinetic parameter estimates in most instances, although at the cost of increased noise compared to the PVC methods utilizing conventional OSEM. Use of the last frame RSFs for PVC with no PSF modelling in the OSEM algorithm produced the lowest bias in cerebral metabolic rate of glucose estimates, although by less than 5% in most cases compared to the other PVC methods. The results indicate that the PVC implementation and choice of PSF modelling in the reconstruction can significantly impact model parameters.
Athinoula A Martinos Center for Biomedical Imaging Bldg 149, Rm 2301, 13th St., Charlestown, MA 02129, USA.
Spencer L
SL
Bowen
Larry G
LG
Byars
Christian J
CJ
Michel
Daniel B
DB
Chonde
Ciprian
C
Catana
D000465:Algorithms; D001921:Brain / Q000033:anatomy & histology* / Q000000981:diagnostic imaging*; D019788:Fluorodeoxyglucose F18; D064368:Healthy Volunteers; D006801:Humans; D007091:Image Processing, Computer-Assisted / Q000379:methods*; D007700:Kinetics; D008954:Models, Biological; D009010:Monte Carlo Method; D009929:Organ Size; D019047:Phantoms, Imaging; D049268:Positron-Emission Tomography; D059629:Signal-To-Noise Ratio
D019788:Fluorodeoxyglucose F18