Development and evaluation of a LOR-based image reconstruction with 3D system response modeling for a PET insert with dual-layer offset crystal design.
In this study we present a method of 3D system response calculation for analytical computer simulation and statistical image reconstruction for a magnetic resonance imaging (MRI) compatible positron emission tomography (PET) insert system that uses a dual-layer offset (DLO) crystal design. The general analytical system response functions (SRFs) for detector geometric and inter-crystal penetration of coincident crystal pairs are derived first. We implemented a 3D ray-tracing algorithm with 4π sampling for calculating the SRFs of coincident pairs of individual DLO crystals. The determination of which detector blocks are intersected by a gamma ray is made by calculating the intersection of the ray with virtual cylinders with radii just inside the inner surface and just outside the outer-edge of each crystal layer of the detector ring. For efficient ray-tracing computation, the detector block and ray to be traced are then rotated so that the crystals are aligned along the X-axis, facilitating calculation of ray/crystal boundary intersection points. This algorithm can be applied to any system geometry using either single-layer (SL) or multi-layer array design with or without offset crystals. For effective data organization, a direct lines of response (LOR)-based indexed histogram-mode method is also presented in this work. SRF calculation is performed on-the-fly in both forward and back projection procedures during each iteration of image reconstruction, with acceleration through use of eight-fold geometric symmetry and multi-threaded parallel computation. To validate the proposed methods, we performed a series of analytical and Monte Carlo computer simulations for different system geometry and detector designs. The full-width-at-half-maximum of the numerical SRFs in both radial and tangential directions are calculated and compared for various system designs. By inspecting the sinograms obtained for different detector geometries, it can be seen that the DLO crystal design can provide better sampling density than SL or dual-layer no-offset system designs with the same total crystal length. The results of the image reconstruction with SRFs modeling for phantom studies exhibit promising image recovery capability for crystal widths of 1.27-1.43 mm and top/bottom layer lengths of 4/6 mm. In conclusion, we have developed efficient algorithms for system response modeling of our proposed PET insert with DLO crystal arrays. This provides an effective method for both 3D computer simulation and quantitative image reconstruction, and will aid in the optimization of our PET insert system with various crystal designs.
Department of Radiology, University of Manitoba, Winnipeg, Manitoba, Canada. Department of Biomedical Engineering, University of California, Davis, Davis, CA, USA.
Xuezhu
X
Zhang
Greg
G
Stortz
Vesna
V
Sossi
Christopher J
CJ
Thompson
Fabrice
F
Retière
Piotr
P
Kozlowski
Jonathan D
JD
Thiessen
Andrew L
AL
Goertzen
D003198:Computer Simulation; D004867:Equipment Design; D021621:Imaging, Three-Dimensional / Q000379:methods*; D008279:Magnetic Resonance Imaging; D009010:Monte Carlo Method; D019047:Phantoms, Imaging; D049268:Positron-Emission Tomography / Q000295:instrumentation*