title: Damage of stone baskets by endourologic lithotripters: a laboratory study of 5 lithotripters and 4 basket types.
abstract: Background. In some cases, the ureteral stone is simultaneously stabilized by a stone basket when endourologic lithotripsy is performed. This stabilization can be either on purpose or by accident. By accident means that an impaction in the ureter occurs by an extraction of a stone with a basket. A stabilization on purpose means to avoid a retropulsion of the stone into the kidney during lithotripsy. At this part of the operation, stone baskets have been frequently damaged. This severing of wires can lead to ureteral trauma because of hook formation. Material and Methods. In a laboratory setting, the time and the pulse numbers were measured until breaking the wires from four different nitinol stone baskets by using five different lithotripsy devices. The endpoint was gross visibledamage to the wire and loss of electric conduction. Results. The Ho:YAG laser and the ultrasonic device were able to destroy almost all the wires. The ballistic devices and the electrohydraulic device were able to destroy thin wires. Conclusion. The operating surgeon should know the risk of damagefor every lithotripter. The Ho:YAG-laser and the ultrasonic device should be classified as dangerous for the basket wire with all adverse effects to the patient.
authors.affiliation: Clinic of Urology, Luebeck Campus, University Medical Center Schleswig-Holstein, Ratzeburger Allee 160, 23538 Luebeck, Germany.
authors.forename: Jens
authors.initials: J
authors.lastname: Cordes
authors.forename: Felix
authors.initials: F
authors.lastname: Nguyen
authors.forename: Birgit
authors.initials: B
authors.lastname: Lange
authors.forename: Ralf
authors.initials: R
authors.lastname: Brinkmann
authors.forename: Dieter
authors.initials: D
authors.lastname: Jocham