Phantom for assessment of fat suppression in large field-of-view diffusion-weighted magnetic resonance imaging.
We present the development and application of a phantom for assessment and optimization of fat suppression over a large field-of-view in diffusion-weighted magnetic resonance imaging at 1.5 T and 3 T. A Perspex cylinder (inner diameter 185 mm, height 300 mm) which contains a second cylinder (inner diameter 140 mm) was constructed. The inner cylinder was filled with water doped with copper sulphate and sodium chloride and the annulus was filled with corn oil, which closely matches the spectrum and longitudinal relaxation times of subcutaneous abdominal fat. Placement of the phantom on the couch at 45° to the z-axis presented an elliptical cross-section, which was of a similar size and shape to axial abdominal images. The use of a phantom for optimization of fat suppression allowed quantitative comparison between studies without the differences introduced by variability between human subjects. We have demonstrated that the phantom is suitable for selection of inversion delay times, spectral adiabatic inversion recovery delays and assessment of combinatorial methods of fat suppression. The phantom is valuable in protocol development and the assessment of new techniques, particularly in multi-centre trials.
CRUK and EPSRC Cancer Imaging Centre, Division of Radiotherapy and Imaging, Institute of Cancer Research, 15 Cotswold Road, Sutton, Surrey, SM2 5NG, UK.
J M
JM
Winfield
N H M
NH
Douglas
N M
NM
deSouza
D J
DJ
Collins
D038524:Diffusion Magnetic Resonance Imaging / Q000295:instrumentation*; D005260:Female; D006801:Humans; D007091:Image Processing, Computer-Assisted; D008297:Male; D019047:Phantoms, Imaging