On using the dosimetric leaf gap to model the rounded leaf ends in VMAT/RapidArc plans.
Partial transmission through rounded leaf ends of Varian multileaf collimators (MLC) is accounted for with a parameter called the dosimetric leaf gap (DLG). Verification of the value of the DLG is needed when the dose delivery is accompanied by gantry rotation in VMAT plans. We compared the doses measured with GAFCHROMIC film and an ionization chamber to treatment planning system (TPS) calculations to identify the optimum values of the DLG in clinical plans of the whole brain with metastases transferred to a phantom. We noticed the absence of a single value of the DLG that properly models all VMAT plans in our cohort (the optimum DLG varied between 0.93 ± 0.15 mm and 2.2 ± 0.2 mm). The former value is considerably different from the optimum DLG in sliding window plans (about 2.0 mm) that approximate IMRT plans. We further found that a single-value DLG model cannot accurately reproduce the measured dose profile even of a uniform static slit at a fixed gantry, which is the simplest MLC-delimited field. The calculation overestimates the measurement in the proximal penumbra, while it underestimates in the distal penumbra. This prompted us to expand the DLG parameter from a plan-specific number to a mathematical concept of the DLG being a function of the distance in the beam's eye view (BEV) between the dose point and the leaf ends. Such function compensates for the difference between the penumbras in a beam delimited with a rounded leaf MLC and delimited with solid jaws. Utilization of this concept allowed us generating a pair of step-and-shoot MLC plans for which we could qualitatively predict the value of the DLG providing best match to ionization chamber measurements. The plan for which the leafs stayed predominantly at positions requiring low values of the DLG (as seen in the profiles of 1D slits) yielded the combined DLG of 1.1 ± 0.2 mm, while the plan with leafs staying at positions requiring larger values of the DLG yielded the DLG 2.4 ± 0.2 mm. Considering the DLG to be a function of the distance (in BEV) between the dose point and the leaf ends allowed us to provide an explanation as to why conventional single-number DLG is plan-specific in VMAT plans.
BC Cancer Agency (Fraser Valley Centre). sszpala@bccancer.bc.ca.
Stanislaw
S
Szpala
Fred
F
Cao
Kirpal
K
Kohli
D000465:Algorithms; D001932:Brain Neoplasms / Q000000981:diagnostic imaging / Q000532:radiotherapy; D015331:Cohort Studies; D005373:Film Dosimetry / Q000295:instrumentation / Q000379:methods*; D006801:Humans; D009362:Neoplasm Metastasis; D011859:Radiography; D011874:Radiometry / Q000295:instrumentation / Q000379:methods*; D011879:Radiotherapy Dosage; D011880:Radiotherapy Planning, Computer-Assisted / Q000379:methods*; D050397:Radiotherapy, Intensity-Modulated / Q000295:instrumentation / Q000379:methods*