Membrane oxygenator heat exchanger failure detected by unique blood gas findings.
Failure of components integrated into the cardiopulmonary bypass circuit, although rare, can bring about catastrophic results. One of these components is the heat exchanger of the membrane oxygenator. In this compartment, unsterile water from the heater cooler device is separated from the sterile blood by stainless steel, aluminum, or by polyurethane. These areas are glued or welded to keep the two compartments separate, maintaining sterility of the blood. Although quality control testing is performed by the manufacturer at the factory level, transport presents the real possibility for damage. Because of this, each manufacturer has included in the instructions for use a testing procedure for testing the integrity of the heat exchanger component. Water is circulated through the heat exchanger before priming and a visible check is made of the oxygenator bundle to check for leaks. If none are apparent, then priming of the oxygenator is performed. In this particular case, this procedure was not useful in detecting communication between the water and blood chambers of the oxygenator.
Justin L
JL
Hawkins
D001784:Blood Gas Analysis / Q000295:instrumentation* / Q000379:methods*; D004866:Equipment Contamination / Q000517:prevention & control*; D004868:Equipment Failure; D019544:Equipment Failure Analysis / Q000295:instrumentation* / Q000379:methods*; D004869:Equipment Safety / Q000295:instrumentation / Q000379:methods; D015199:Extracorporeal Membrane Oxygenation / Q000295:instrumentation*; D006361:Heating / Q000295:instrumentation*; D006801:Humans; D007223:Infant