Fully automated laser ablation liquid capture surface analysis using nanoelectrospray ionization mass spectrometry.
BACKGROUND
Laser ablation provides for the possibility of sampling a large variety of surfaces with high spatial resolution. This type of sampling when employed in conjunction with liquid capture followed by nanoelectrospray ionization provides the opportunity for sensitive and prolonged interrogation of samples by mass spectrometry as well as the ability to analyze surfaces not amenable to direct liquid extraction.


METHODS
A fully automated, reflection geometry, laser ablation liquid capture spot sampling system was achieved by incorporating appropriate laser fiber optics and a focusing lens into a commercially available, liquid extraction surface analysis (LESA(®))-ready Advion TriVersa NanoMate system.


RESULTS
Under optimized conditions about 10% of laser-ablated material could be captured in a droplet positioned vertically over the ablation region using the NanoMate robot-controlled pipette. The sampling spot size area with this laser ablation liquid capture surface analysis (LA/LCSA) mode of operation (typically about 120 µm × 160 µm) was approximately 50 times smaller than that achievable by direct liquid extraction using LESA(®) (ca 1 mm diameter liquid extraction spot). The setup was successfully applied for the analysis of ink on glass and paper as well as the endogenous components in Alstroemeria Yellow King flower petals. In a second mode of operation with a comparable sampling spot size, termed laser ablation/LESA(®), the laser system was used to drill through, penetrate, or otherwise expose material beneath a solvent resistant surface. Once drilled, LESA(®) was effective in sampling soluble material exposed at that location on the surface.


CONCLUSIONS
Incorporating the capability for different laser ablation liquid capture spot sampling modes of operation into a LESA(®)-ready Advion TriVersa NanoMate enhanced the spot sampling spatial resolution of this device and broadened the surface types amenable to analysis to include absorbent and solvent-resistant materials.
Organic and Biological Mass Spectrometry Group, Chemical Sciences Division, Oak Ridge National Laboratory, Oak Ridge, TN 37831-6131, USA.
Matthias
M
Lorenz
Olga S
OS
Ovchinnikova
Gary J
GJ
Van Berkel