title: Design and performance evaluation of a microfluidic ion-suppression module for anion-exchange chromatography.
abstract: A microfluidic membrane suppressor has been constructed to suppress ions of alkaline mobile-phases via an acid-base reaction across a sulfonated poly(tetrafluoroethylene)-based membrane and was evaluated for anion-exchange separations using conductivity detection. The membrane was clamped between two chip substrates, accommodating rectangular microchannels for the eluent and regenerant flow, respectively. Additionally, a clamp-on chip holder has been constructed which allows the alignment and stacking of different chip modules. The response and efficacy of the microfluidic chip suppressor was assessed for a wide range of eluent (KOH) concentrations, using 127 and 183μm thick membranes, while optimizing the flow rate and concentration of the regenerant solution (H2SO4). The optimal operating eluent flow rate was determined at 5μL/min, corresponding to the optimal van-Deemter flow velocity of commercially-available column technology, i.e. a 0.4mm i.d.×250mm long column packed with 7.5μm anion-exchange particles. When equilibrated at 10mM KOH, a 99% decrease in conductivity signal could be obtained within 5min when applying 10mM H2SO4 regenerant at 75μL/min. A background signal as low as 1.2μS/cm was obtained, which equals the performance of a commercially-available electrolytic hollow-fiber suppressor. When increasing the temperature of the membrane suppressor from 15 to 20°C, ion suppression was significantly improved allowing the application of 75mM KOH. The applicability of the chip suppressor has been demonstrated with an isocratic baseline separation of a mixture of seven inorganic ions, yielding plate numbers between 5300 and 10,600 and with a gradient separation of a complex ion mixture.
authors: Vrije Universiteit Brussel, Department of Chemical Engineering, Pleinlaan 2, B-1050 Brussels, Belgium.
authors: Sam
authors: S
authors: Wouters
authors: Vrije Universiteit Brussel, Department of Chemical Engineering, Pleinlaan 2, B-1050 Brussels, Belgium.
authors: Bert
authors: B
authors: Wouters
authors: Vrije Universiteit Brussel, Department of Chemical Engineering, Pleinlaan 2, B-1050 Brussels, Belgium.
authors: Sander
authors: S
authors: Jespers
authors: Vrije Universiteit Brussel, Department of Chemical Engineering, Pleinlaan 2, B-1050 Brussels, Belgium.
authors: Gert
authors: G
authors: Desmet
authors: Vrije Universiteit Brussel, Department of Chemical Engineering, Pleinlaan 2, B-1050 Brussels, Belgium.
authors: Hamed
authors: H
authors: Eghbali
authors: Leiden University Medical Center, Center for Proteomics and Metabolomics, PO box 9600, 2300 RC Leiden, The Netherlands.
authors: Cees
authors: C
authors: Bruggink
authors: Vrije Universiteit Brussel, Department of Chemical Engineering, Pleinlaan 2, B-1050 Brussels, Belgium. Electronic address: seeltink@vub.ac.be.
authors: Sebastiaan
authors: S
authors: Eeltink
mesh_terms: D000838:Anions / Q000737:chemistry; D002852:Chromatography, Ion Exchange / Q000295:instrumentation / Q000379:methods*; D004572:Electrolysis; D044085:Microfluidics / Q000295:instrumentation / Q000379:methods*
chemical_list: D000838:Anions
keywords: Conductivity detection; Ion chromatography; Membrane suppressor; Miniaturization