title: Dual-element transducer with phase-inversion for wide depth of field in high-frequency ultrasound imaging.
abstract: In high frequency ultrasound imaging (HFUI), the quality of focusing is deeply related to the length of the depth of field (DOF). In this paper, a phase-inversion technique implemented by a dual-element transducer is proposed to enlarge the DOF. The performance of the proposed method was numerically demonstrated by using the ultrasound simulation program called Field-II. A simulated dual-element transducer was composed of a disc- and an annular-type elements, and its aperture was concavely shaped to have a confocal point at 6 mm. The area of each element was identical in order to provide same intensity at the focal point. The outer diameters of the inner and the outer elements were 2.1 mm and 3 mm, respectively. The center frequency of each element was 40 MHz and the f-number (focal depth/aperture size) was two. When two input signals with 0° and 180° phases were applied to inner and outer elements simultaneously, a multi-focal zone was generated in the axial direction. The total -6 dB DOF, i.e., sum of two -6 dB DOFs in the near and far field lobes, was 40% longer than that of the conventional single element transducer. The signal to noise ratio (SNR) was increased by about two times, especially in the far field. The point and cyst phantom simulation were conducted and their results were identical to that of the beam pattern simulation. Thus, the proposed scheme may be a potential method to improve the DOF and SNR in HFUI.
authors.affiliation: Department of Medical Biotechnology, Dongguk University, Seoul 100-715, Korea. jjsspace@dongguk.edu.
authors.forename: Jong Seob
authors.initials: JS
authors.lastname: Jeong
mesh_terms: D004867:Equipment Design / Q000295:instrumentation / Q000379:methods; D019047:Phantoms, Imaging; D059629:Signal-To-Noise Ratio; D014159:Transducers; D014465:Ultrasonics / Q000295:instrumentation* / Q000379:methods*
references.citation: J Acoust Soc Am. 1991 Jan;89(1):182-90
references.pmid: 2002167
references.citation: Appl Opt. 2002 Sep 1;41(25):5263-6
references.pmid: 12211552
references.citation: IEEE Trans Ultrason Ferroelectr Freq Control. 2006 Mar;53(3):623-30
references.pmid: 16555771
references.citation: IEEE Trans Ultrason Ferroelectr Freq Control. 2005 Apr;52(4):672-81
references.pmid: 16060516
references.citation: IEEE Trans Ultrason Ferroelectr Freq Control. 1989;36(5):540-8
references.pmid: 18290231
references.citation: IEEE Trans Biomed Eng. 2008 Aug;55(8):2039-49
references.pmid: 18632366
references.citation: Appl Opt. 2001 Nov 1;40(31):5658-62
references.pmid: 18364855
references.citation: Ophthalmology. 2004 Jul;111(7):1344-51
references.pmid: 15234135
references.citation: IEEE Trans Ultrason Ferroelectr Freq Control. 2008 Feb;55(2):508-13
references.pmid: 18334358
references.citation: IEEE Trans Ultrason Ferroelectr Freq Control. 2005 Mar;52(3):375-85
references.pmid: 15857046
references.citation: IEEE Trans Ultrason Ferroelectr Freq Control. 1992;39(2):262-7
references.pmid: 18263145