title: [Performance of wavelength modulation surface plasmon resonance biosensor].
abstract: Surface plasmon resonance (SPR) is a rapid, label-free, high-precision technique of biological sensing and analysis. The investigation on the characteristics of provides theoretical basis and instructions for the applications of SPR A Kretschmann-structure surface plasmon resonance (SPR) biosensor based on wavelength modulation was developed, and also its sensing performances in the bulk solution was investigated. Measurements with different concentrations of bulk ethanol and ethylene glycol solutions show that the resonant wavelength shows a low sensitivity, but a higher linear response to the change in refractive index (RI), when RI is relatively smaller. With increasing refractive index , the sensitivity of resonance wavelength to changes in the refractive index increases. In the refractive index range of 1. 407 0-1. 430 RIU, sensitivity reaches to 11 487 nm RIU-1. The sensor resonance wavelength stability is 0. 213 8 nm, and the minimum resolution of refractive index approaches to 10-6 RIU. The advantages of the surface plasmon resonance sensor developed here results in simple operation, high sensitivity, wide detection range, low resolution, makes it an important candidate in chemical and biological sensing.
authors.forename: Yun-Han
authors.initials: YH
authors.lastname: Luo
authors.forename: Meng-Yun
authors.initials: MY
authors.lastname: Xu
authors.forename: Xiao-Long
authors.initials: XL
authors.lastname: Chen
authors.forename: Jie-Yuan
authors.initials: JY
authors.lastname: Tang
authors.forename: Fang
authors.initials: F
authors.lastname: Wang
authors.forename: Yi-Long
authors.initials: YL
authors.lastname: Zhang
authors.forename: Yong-Hong
authors.initials: YH
authors.lastname: He
authors.forename: Zhe
authors.initials: Z
authors.lastname: Chen
mesh_terms: D015374:Biosensing Techniques; D012031:Refractometry; D020349:Surface Plasmon Resonance