title: An in vitro biomechanical comparison of hydroxyapatite coated and uncoated ao cortical bone screws for a limited contact: dynamic compression plate fixation of osteotomized equine 3rd metacarpal bones.
abstract: OBJECTIVE
  To compare the monotonic biomechanical properties of a broad 4.5 mm limited contact-dynamic compression plate (LC-DCP) fixation secured with hydroxyapatite (HA) coated cortical bone screws (HA-LC-DCP) versus uncoated cortical bone screws (AO-LC-DCP) to repair osteotomized equine 3rd metacarpal (MC3) bones.
  
  
  METHODS
  Experimental.
  
  
  METHODS
  Adult equine cadaveric MC3 bones (n = 12 pair).
  
  
  METHODS
  Twelve pairs of equine MC3 were divided into 3 test groups (4 pairs each) for: (1) 4 point bending single cycle to failure testing; (2) 4 point bending cyclic fatigue testing; and (3) torsional single cycle to failure testing. For the HA-LC-DCP-MC3 construct, an 8-hole broad LC-DCP (Synthes Ltd, Paoli, PA) was secured on the dorsal surface of each randomly selected MC3 bone with a combination of four 5.5 mm and four 4.5 mm HA-coated cortical screws. For the AO-LC-DCP-MC3 construct, an 8-hole 4.5 mm broad LC-DCP was secured on the dorsal surface of the contralateral MC3 bone with a combination of four 5.5 mm and four 4.5 mm uncoated cortical screws. All MC3 bones had mid-diaphyseal osteotomies. Mean test variable values for each method were compared using a paired t-test within each group. Significance was set at P < .05.
  
  
  RESULTS
  Mean yield load, yield bending moment, composite rigidity, failure load, and failure bending moment, under 4 point bending, single cycle to failure, of the HA-LC-DCP fixation were significantly greater than those of the AO-LC-DCP fixation. Mean ± SD values for the HA-LC-DCP and the AO-LC-DCP fixation techniques, respectively, in single cycle to failure under 4 point bending were: yield load, 26.7 ± 2.15 and 16.3 ± 1.38 kN; yield bending moment, 527.4 ± 42.4 and 322.9 ± 27.2 N-m; composite rigidity, 5306 ± 399 and 3003 ± 300 N-m/rad; failure load, 40.6 ± 3.94 and 26.5 ± 2.52 kN; and failure bending moment, 801.9 ± 77.9 and 522.9 ± 52.2 N-m. Mean cycles to failure in 4 point bending of the HA-LC-DCP fixation (116,274 ± 13,211) was significantly greater than that of the AO-LC-DCP fixation 47,619 ± 6580. Mean yield load, mean composite rigidity, and mean failure load under torsional testing, single cycle to failure was significantly greater for the broad HA-LC-DCP fixation compared with the AO-LC-DCP fixation. In single cycle to failure under torsion, mean ± SD values for the HA-LC-DCP and the AO-LC-DCP fixation techniques, respectively, were: yield load, 101.3 ± 14.68 and 70.54 ± 10.20 N-m; composite rigidity, 437.9 ± 32.9 and 220.7 ± 17.6 N-m/rad; and failure load: 105.7 ± 15.5 and 75.28 ± 10.1 N-m.
  
  
  CONCLUSIONS
  HA-LC-DCP was superior to AO-LC-DCP in resisting the static overload forces (palmarodorsal 4 point bending and torsional) and in resisting cyclic fatigue under palmarodorsal 4 point bending.
authors.affiliation: Equine Health Studies Program, Department of Veterinary Clinical Sciences, School of Veterinary Medicine, Louisiana State University, Baton Rouge, Louisiana.
authors.forename: Myra E
authors.initials: ME
authors.lastname: Durham
authors.forename: Gary A
authors.initials: GA
authors.lastname: Sod
authors.forename: Laura M
authors.initials: LM
authors.lastname: Riggs
authors.forename: Colin F
authors.initials: CF
authors.lastname: Mitchell
mesh_terms: D000818:Animals; D001672:Biocompatible Materials; D001696:Biomechanical Phenomena; D001860:Bone Plates / Q000662:veterinary*; D002102:Cadaver; D017886:Durapatite; D005593:Fracture Fixation, Internal / Q000662:veterinary*; D006736:Horses / Q000293:injuries* / Q000601:surgery; D066298:In Vitro Techniques; D050279:Metacarpal Bones / Q000293:injuries* / Q000601:surgery; D010027:Osteotomy / Q000662:veterinary
chemical_list: D001672:Biocompatible Materials; D017886:Durapatite