Absorbed dose measurements for kV-cone beam computed tomography in image-guided radiation therapy.
In this study, we develope a novel method to directly evaluate an absorbed dose-to-water for kilovoltage-cone beam computed tomography (kV-CBCT) in image-guided radiation therapy (IGRT). Absorbed doses for the kV-CBCT systems of the Varian On-Board Imager (OBI) and the Elekta X-ray Volumetric Imager (XVI) were measured by a Farmer ionization chamber with a (60)Co calibration factor. The chamber measurements were performed at the center and four peripheral points in body-type (30 cm diameter and 51 cm length) and head-type (16 cm diameter and 33 cm length) cylindrical water phantoms. The measured ionization was converted to the absorbed dose-to-water by using a (60)Co calibration factor and a Monte Carlo (MC)-calculated beam quality conversion factor, kQ, for (60)Co to kV-CBCT. The irradiation for OBI and XVI was performed with pelvis and head modes for the body- and the head-type phantoms, respectively. In addition, the dose distributions in the phantom for both kV-CBCT systems were calculated with MC method and were compared with measured values. The MC-calculated doses were calibrated at the center in the water phantom and compared with measured doses at four peripheral points. The measured absorbed doses at the center in the body-type phantom were 1.96 cGy for OBI and 0.83 cGy for XVI. The peripheral doses were 2.36-2.90 cGy for OBI and 0.83-1.06 cGy for XVI. The doses for XVI were lower up to approximately one-third of those for OBI. Similarly, the measured doses at the center in the head-type phantom were 0.48 cGy for OBI and 0.21 cGy for XVI. The peripheral doses were 0.26-0.66 cGy for OBI and 0.16-0.30 cGy for XVI. The calculated peripheral doses agreed within 3% in the pelvis mode and within 4% in the head mode with measured doses for both kV-CBCT systems. In addition, the absorbed dose determined in this study was approximately 4% lower than that in TG-61 but the absorbed dose by both methods was in agreement within their combined uncertainty. This method is more robust and accurate compared to the dosimetry based on a conventional air-kerma calibration factor. Therefore, it is possible to be used as a standard dosimetry protocol for kV-CBCT in IGRT.
Section of Radiation Therapy, Department of Clinical Support, Hiroshima University Hospital, 1-2-3 Kasumi, Hiroshima, 734-8551, Japan.
Kazunari
K
Hioki
Fujio
F
Araki
Takeshi
T
Ohno
Yuji
Y
Nakaguchi
Yuuki
Y
Tomiyama
D000465:Algorithms; D002138:Calibration; D054893:Cone-Beam Computed Tomography; D019047:Phantoms, Imaging; D011829:Radiation Dosage; D011874:Radiometry / Q000379:methods* / Q000592:standards; D061089:Radiotherapy, Image-Guided