title: Partial airway obstruction following manufacturing defect in laryngeal mask airway (Laryngeal Mask Silken™).
abstract: Laryngeal mask (LM) airway is commonly used for securing airway in day-care surgeries. Various problems have been described while using LM airway. Out of those, mechanical obstruction causing airway compromise is most common. Here, we describe a case report of 4-year-old child who had partial upper airway obstruction due to LM manufacturer's defect. There was a silicon band in upper one-third of shaft of LM airway. This band was made up of the same material as that of LM airway so it was not identifiable on external inspection of transparent shaft. We suggest that such as non-transparent laryngeal mask, a transparent LM airway should also be inspected looking inside the lumen with naked eyes or by using a probe to rule out any manufacturing defect before its insertion.
authors.affiliation: Department of Anesthesia and Intensive Care, Post-Graduate Institute of Medical Education and Research, Chandigarh, India.
authors.forename: Kiran
authors.initials: K
authors.lastname: Jangra
authors.affiliation: Department of Anesthesia and Intensive Care, Post-Graduate Institute of Medical Education and Research, Chandigarh, India.
authors.forename: Surender Kumar
authors.initials: SK
authors.lastname: Malhotra
authors.affiliation: Department of Anesthesia and Intensive Care, Post-Graduate Institute of Medical Education and Research, Chandigarh, India.
authors.forename: Vikas
authors.initials: V
authors.lastname: Saini
keywords: Airway; laryngeal mask airway; obstruction