title: A Simple Capacitive Charge-Division Readout for Position-Sensitive Solid-State Photomultiplier Arrays.
abstract: A capacitive charge-division readout method for reading out a 2 × 2 array of 5 mm × 5 mm position-sensitive solid-state photomultipliers (PS-SSPM) was designed and evaluated. Using this analog multiplexing method, the 20 signals (16 position, 4 timing) from the PS-SSPM array are reduced to 5 signals (4 position, 1 timing), allowing the PS-SSPM array to be treated as an individual large-area PS-SSPM module. A global positioning approach can now be used, instead of individual positioning for each PS-SSPM in the array, ensuring that the entire light signal is utilized. The signal-to-noise ratio (SNR) and flood histogram quality at different bias voltages (27.5 V to 32.0 V at 0.5 V intervals) and a fixed temperature of 0 °C were evaluated by coupling a 6 × 6 array of 1.3 mm × 1.3 mm × 20 mm polished LSO crystals to the center of the PS-SSPM array. The timing resolution was measured at a fixed bias voltage of 31.0 V and a fixed temperature of 0 °C. All the measurements were evaluated and compared using capacitors with different values and tolerances. Capacitor values ranged from 0.051 nf to 10 nf, and the capacitance tolerance ranged from 1% to 20%. The results show that better performance was achieved using capacitors with smaller values and better capacitance tolerance. Using 0.2 nf capacitors, the SNR, energy resolution and timing resolution were 24.3, 18.2% and 8.8 ns at a bias voltage 31.0 V, respectively. The flood histogram quality was also evaluated by using a 10 × 10 array of 1 mm × 1 mm × 10 mm polished LSO crystals and a 10 × 10 array of 0.7 mm × 0.7 mm × 20 mm unpolished LSO crystals to determine the smallest crystal size resolvable. These studies showed that the high spatial resolution of the PS-SSPM was preserved allowing for 0.7 mm crystals to be identified. These results show that the capacitive charge-division analog signal processing method can significantly reduce the number of electronic channels, from 20 to 5, while retaining the excellent performance of the detector.
authors.affiliation: Department of Biomedical Engineering, University of California, Davis, CA 95616, USA.
authors.forename: Junwei
authors.initials: J
authors.lastname: Du
authors.affiliation: Department of Biomedical Engineering, University of California, Davis, CA 95616, USA.
authors.forename: Jeffrey P
authors.initials: JP
authors.lastname: Schmall
authors.affiliation: Department of Biomedical Engineering, University of California, Davis, CA 95616, USA.
authors.forename: Yongfeng
authors.initials: Y
authors.lastname: Yang
authors.affiliation: Department of Biomedical Engineering, University of California, Davis, CA 95616, USA.
authors.forename: Kun
authors.initials: K
authors.lastname: Di
authors.affiliation: Radiation Monitoring Devices Inc., Watertown, MA 02172, USA.
authors.forename: Purushottam A
authors.initials: PA
authors.lastname: Dokhale
authors.affiliation: Radiation Monitoring Devices Inc., Watertown, MA 02172, USA.
authors.forename: Kanai S
authors.initials: KS
authors.lastname: Shah
authors.affiliation: Department of Biomedical Engineering, University of California, Davis, CA 95616, USA.
authors.forename: Simon R
authors.initials: SR
authors.lastname: Cherry
keywords: Capacitive Charge–Division Readout; Electronics; PET; PS-SSPM