title: Evaluation of Laplacian diaphragm electromyographic recording in a dynamic inspiratory maneuver.
abstract: The analysis of the electromyographic signal of the diaphragm muscle (EMGdi) can provide important information for evaluating the respiratory muscular function. The EMGdi can be recorded using surface Ag/AgCl disc electrodes in monopolar or bipolar configuration. However, these non-invasive EMGdi recordings are usually contaminated by the electrocardiographic (ECG) signal. EMGdi signal can also be noninvasively recorded using concentric ring electrodes in bipolar configuration (CRE) that estimate Laplacian surface potential. Laplacian recordings increase spatial resolution and attenuate distant bioelectric interferences, such as the ECG. Thus, the objective of this work is to compare and to evaluate CRE and traditional bipolar EMGdi recordings in a healthy subject during a dynamic inspiratory maneuver with incremental inspiratory loads. In the conducted study, it was calculated the cumulative percentage of power spectrum of EMGdi recordings to determine the signal bandwidth, and the power ratio between the EMGdi signal segments with and without cardiac activity. The results of this study suggest that EMGdi acquired with CRE electrodes is less affected by the ECG interference, achieves a wider bandwidth and a higher power ratio between segments without cardiac activity and with cardiac activity.
authors.forename: Luis
authors.initials: L
authors.lastname: Estrada
authors.forename: Abel
authors.initials: A
authors.lastname: Torres
authors.forename: Javier
authors.initials: J
authors.lastname: Garcia-Casado
authors.forename: Gema
authors.initials: G
authors.lastname: Prats-Boluda
authors.lastname: Yiyao Ye-Lin
authors.forename: Raimon
authors.initials: R
authors.lastname: Jane
mesh_terms: D003964:Diaphragm / Q000502:physiology*; D004562:Electrocardiography; D004576:Electromyography / Q000379:methods*; D006801:Humans; D001239:Inhalation; D008297:Male; D012815:Signal Processing, Computer-Assisted; D013990:Tidal Volume