title: Analysis of rapid multi-focal-zone ARFI imaging.
abstract: Acoustic radiation force impulse (ARFI) imaging has shown promise for visualizing structure and pathology within multiple organs; however, because the contrast depends on the push beam excitation width, image quality suffers outside of the region of excitation. Multi-focal-zone ARFI imaging has previously been used to extend the region of excitation (ROE), but the increased acquisition duration and acoustic exposure have limited its utility. Supersonic shear wave imaging has previously demonstrated that through technological improvements in ultrasound scanners and power supplies, it is possible to rapidly push at multiple locations before tracking displacements, facilitating extended depth of field shear wave sources. Similarly, ARFI imaging can utilize these same radiation force excitations to achieve tight pushing beams with a large depth of field. Finite element method simulations and experimental data are presented, demonstrating that single- and rapid multi-focal-zone ARFI have comparable image quality (less than 20% loss in contrast), but the multi-focal-zone approach has an extended axial region of excitation. Additionally, as compared with single-push sequences, the rapid multi-focalzone acquisitions improve the contrast-to-noise ratio by up to 40% in an example 4-mm-diameter lesion.
authors: Stephen
authors: S
authors: Rosenzweig
authors: Mark
authors: M
authors: Palmeri
authors: Kathryn
authors: K
authors: Nightingale
mesh_terms: D000465:Algorithms; D054459:Elasticity Imaging Techniques / Q000379:methods*; D020342:Finite Element Analysis; D007091:Image Processing, Computer-Assisted / Q000379:methods*; D019047:Phantoms, Imaging; D059629:Signal-To-Noise Ratio
references: IEEE Trans Ultrason Ferroelectr Freq Control. 1992;39(2):262-7
references: 18263145
references: IEEE Trans Ultrason Ferroelectr Freq Control. 2005 Oct;52(10):1699-712
references: 16382621
references: Ultrasound Med Biol. 2008 Sep;34(9):1373-86
references: 18395961
references: Heart Rhythm. 2012 Nov;9(11):1855-62
references: 22772134
references: Ultrasound Med Biol. 1998 Nov;24(9):1419-35
references: 10385964
references: Ultrason Imaging. 2009 Jul;31(3):172-82
references: 19771960
references: Ultrasound Med Biol. 2009 May;35(5):707-16
references: 19243877
references: IEEE Trans Image Process. 1992;1(2):148-61
references: 18296150
references: IEEE Trans Ultrason Ferroelectr Freq Control. 2013 Nov;60(11):2347-58
references: 24158290
references: Ultrasound Med Biol. 2012 Jan;38(1):50-61
references: 22104533
references: J Acoust Soc Am. 2001 Jul;110(1):625-34
references: 11508987
references: IEEE Trans Ultrason Ferroelectr Freq Control. 2006 Jun;53(6):1103-17
references: 16846143
references: IEEE Trans Ultrason Ferroelectr Freq Control. 2004 Apr;51(4):396-409
references: 15139541
references: IEEE Trans Ultrason Ferroelectr Freq Control. 2007 Feb;54(2):301-12
references: 17328327
references: Ultrasound Med Biol. 2006 Jan;32(1):61-72
references: 16364798
references: IEEE Trans Ultrason Ferroelectr Freq Control. 2003 Jun;50(6):631-41
references: 12839175
references: IEEE Trans Ultrason Ferroelectr Freq Control. 2006 Jul;53(7):1300-13
references: 16889337
references: Phys Med Biol. 2008 Jan 7;53(1):279-93
references: 18182703
references: IEEE Trans Ultrason Ferroelectr Freq Control. 2011 Feb;58(2):399-405
references: 21342825
references: Ultrasound Med Biol. 2002 Feb;28(2):227-35
references: 11937286