title: Topographic Analysis of HyFlex(®) Controlled Memory Nickel-Titanium Files.
abstract: BACKGROUND
  The multiple uses of HyFlex(®) controlled memory (CM) rotary instruments (Coltene-Whaledent, Allstetten, Switzerland) have been recommended after sterilization. The purpose of this study was to analyze the surface defects on HyFlex(®) instruments after use and to determine their ability to return to their original shape after autoclaving.
  
  
  METHODS
  Twenty-four new HyFlex(®) files were scanned using micro-computed tomography before the initial use, after use until defects were visible and after autoclaving. Surface analysis was performed on each file to determine the changes in the files after deformation; defects were recorded as unwinding and curving. These changes were categorized according to the severity, and the initial and final scans of the files were superimposed to illustrate the recovery of the files to their original shape. The data were analyzed using Chi-square and t-test.
  
  
  RESULTS
  The t-test of Student's Newman-Keuls revealed a significant difference (P < 0.05) between unused and used files, almost all files exhibited visible defects without fractures, including unwinding and curving associated with unwinding. The most frequently observed defect was unwinding of the spirals (100%), whereas curving associated with unwinding was infrequently observed. A Chi-square test showed that the deformation (curve: P = 0.000, and unwinding: P = 0.011, incidence depended on the size of the files. The greatest frequency of curving occurred in the small files #20/0.04 and #20/0.06, the difference was statistically significant from all file sizes (P < 0.05). The majority of the files (79%) were able to restore the unwinding and regain their original shape after autoclaving. The highest rate of unrestored files was observed on #20/0.04 and #20/0.06 instruments, and the difference was statistically significant from all file sizes (P < 0.05).
  
  
  CONCLUSIONS
  HyFlex(®) CM nickel-titanium files appear to regain their original shape after sterilization.
authors: Associate Professor, Department of Restorative Dental Sciences, Division of Endodontics, College of Dentistry, King Saud University, Riyadh, Saudi Arabia.
authors: Dina
authors: D
authors: Al-Sudani
keywords: Controlled memory alloy; HyFlex®; controlled memory; endodontics; nickel titanium alloy
references: J Endod. 2013 Nov;39(11):1444-7
references: 24139271
references: J Endod. 2012 Nov;38(11):1535-40
references: 23063231
references: J Endod. 2008 Aug;34(8):1003-5
references: 18634935
references: J Endod. 2004 Oct;30(10):722-5
references: 15448468
references: Int Endod J. 2011 Apr;44(4):337-41
references: 21219362
references: J Endod. 2001 Aug;27(8):516-20
references: 11501589
references: J Dent Res. 2000 Jun;79(6):1405-9
references: 10890720
references: Aust Endod J. 2015 Apr;41(1):24-8
references: 24697976
references: Int Endod J. 2000 Jul;33(4):297-310
references: 11307203
references: Int Endod J. 2009 Jul;42(7):621-6
references: 19467049
references: Int Endod J. 2001 Jan;34(1):47-53
references: 11307380
references: J Endod. 2002 Oct;28(10):716-20
references: 12398171
references: J Endod. 2013 Mar;39(3):385-8
references: 23402512
references: J Endod. 2013 Jan;39(1):101-4
references: 23228266
references: J Endod. 2013 Feb;39(2):163-72
references: 23321225
references: Int Endod J. 2007 Apr;40(4):247-53
references: 17284269
references: J Endod. 2012 Mar;38(3):376-80
references: 22341078
references: Int Endod J. 2001 Apr;34(3):221-30
references: 12193268
references: Int Endod J. 2003 Nov;36(11):740-7
references: 14641437
references: J Endod. 2011 Jul;37(7):997-1001
references: 21689559
references: J Endod. 2008 Nov;34(11):1406-9
references: 18928858
references: J Endod. 2006 Jan;32(1):61-5
references: 16410072
references: J Endod. 2009 Feb;35(2):193-6
references: 19166771