title: One-heater flow-through polymerase chain reaction device by heat pipes cooling.
abstract: This study describes a novel microfluidic reactor capable of flow-through polymerase chain reactions (PCR). For one-heater PCR devices in previous studies, comprehensive simulations and experiments for the chip geometry and the heater arrangement were usually needed before the fabrication of the device. In order to improve the flexibility of the one-heater PCR device, two heat pipes with one fan are used to create the requisite temperature regions in our device. With the integration of one heater onto the chip, the high temperature required for the denaturation stage can be generated at the chip center. By arranging the heat pipes on the opposite sides of the chip, the low temperature needed for the annealing stage is easy to regulate. Numerical calculations and thermal measurements have shown that the temperature distribution in the five-temperature-region PCR chip would be suitable for DNA amplification. In order to ensure temperature uniformity at specific reaction regions, the Re of the sample flow is less than 1. When the microchannel width increases and then decreases gradually between the denaturation and annealing regions, the extension region located in the enlarged part of the channel can be observed numerically and experimentally. From the simulations, the residence time at the extension region with the enlarged channel is 4.25 times longer than that without an enlarged channel at a flow rate of 2 μl/min. The treated surfaces of the flow-through microchannel are characterized using the water contact angle, while the effects of the hydrophilicity of the treated polydimethylsiloxane (PDMS) microchannels on PCR efficiency are determined using gel electrophoresis. By increasing the hydrophilicity of the channel surface after immersing the PDMS substrates into Tween 20 (20%) or BSA (1 mg/ml) solutions, efficient amplifications of DNA segments were proved to occur in our chip device. To our knowledge, our group is the first to introduce heat pipes into the cooling module that has been designed for a PCR device. The unique architecture utilized in this flow-through PCR device is well applied to a low-cost PCR system.
authors.affiliation: Department of Biomechatronics Engineering, National Pingtung University of Science and Technology , 1, Shuefu Road, Neipu, Pingtung 91201, Taiwan.
authors.forename: Jyh Jian
authors.initials: JJ
authors.lastname: Chen
authors.affiliation: Department of Veterinary Medicine, National Pingtung University of Science and Technology , 1, Shuefu Road, Neipu, Pingtung 91201, Taiwan.
authors.forename: Ming Huei
authors.initials: MH
authors.lastname: Liao
authors.affiliation: Department of Biomechatronics Engineering, National Pingtung University of Science and Technology , 1, Shuefu Road, Neipu, Pingtung 91201, Taiwan.
authors.forename: Kun Tze
authors.initials: KT
authors.lastname: Li
authors.affiliation: Department of Biomechatronics Engineering, National Pingtung University of Science and Technology , 1, Shuefu Road, Neipu, Pingtung 91201, Taiwan.
authors.forename: Chia Ming
authors.initials: CM
authors.lastname: Shen