title: [In-phantom dosimetric measurements as quality control for brachytherapy: System check and constancy check].
abstract: In brachytherapy dosimetric measurements are difficult due to the inherent dose-inhomogenieties. Typically in routine clincal practice only the nominal dose rate is determined for computer controlled afterloading systems. The region of interest lies close to the source when measuring the spatial dose distribution. In this region small errors in the postioning of the detector, and its finite size, lead to large measurement uncertainties that exacerbate the routine dosimetric control of the system in the clinic. The size of the measurement chamber, its energy dependence, and the directional dependence of the measurement apparatus are the factors which have a significant influence on dosimetry. Although ionisation chambers are relatively large, they are employed since similar chambers are commonly found on clincal brachytherapy units. The dose is determined using DIN 6800 [11] since DIN 6809-2 [12], which deals with dosimetry in brachytherapy, is antiquated and is currently in the process of revision. Further information regarding dosimetry for brachytherapy can be found in textbooks [1] and [2]. The measurements for this work were performed with a HDR (High-Dose-Rate) (192)Ir source, type mHDR V2, and a Microselectron Afterloader V2 both from Nucletron/Elekta. In this work two dosimetric procedures are presented which, despite the aforemention difficulties, should assist in performing checks of the proper operation of the system. The first is a system check that measures the dose distribution along a line and is to be performed when first bringing the afterloader into operation, or after significant changes to the system. The other is a dosimetric constancy check, which with little effort can be performed monhtly or weekly. It simultaneously verifies the positioning of the source at two positions, the functionality of the system clock and the automatic re-calculation of the source activity.
authors: Medizinische Physik, Klinik für Strahlenheilkunde, Universitätsklinik Freiburg. Electronic address: michael.kollefrath@uniklinik-freiburg.de.
authors: Michael
authors: M
authors: Kollefrath
authors: Medizinische Physik, Klinik für Strahlenheilkunde, Universitätsklinik Freiburg. Electronic address: gregor.bruggmoser@uniklinik-freiburg.de.
authors: Gregor
authors: G
authors: Bruggmoser
authors: Medizinische Physik, Klinik für Strahlenheilkunde, Universitätsklinik Freiburg. Electronic address: norbert.nanko@uniklinik-freiburg.de.
authors: Norbert
authors: N
authors: Nanko
authors: Medizinische Physik, Klinik für Strahlenheilkunde, Universitätsklinik Freiburg. Electronic address: mark.gainey@uniklinik-freiburg.de.
authors: Mark
authors: M
authors: Gainey
mesh_terms: D000465:Algorithms; D001918:Brachytherapy / Q000592:standards*; D002138:Calibration; D005858:Germany; D019047:Phantoms, Imaging; D017410:Practice Guidelines as Topic; D011786:Quality Control; D011868:Radioisotopes / Q000032:analysis*; D011874:Radiometry / Q000379:methods* / Q000592:standards*; D011879:Radiotherapy Dosage; D015203:Reproducibility of Results; D012680:Sensitivity and Specificity
chemical_list: D011868:Radioisotopes
keywords: Brachytherapy quality assurance; Messung im Festkörperphantom; Positionierungskontrolle Brachytherapie; Qualitätssicherung Brachytherapie; Systemprüfung Afterloader; afterloader system check; brachytherapy positioning-control; solid phantom measurement