Air compressor battery duration with mechanical ventilation in a field anesthesia machine.
Compressed air to power field anesthesia machine ventilators may be supplied by air compressor with battery backup. This study determined the battery duration when the compPAC ventilator's air compressor was powered by NiCd battery to ventilate the Vent Aid Training Test Lung modeling high (HC = 0.100 L/cm H2O) and low (LC = 0.020 L/cm H2O) pulmonary compliance. Target tidal volumes (VT) were 500, 750, and 1,000 mL. Respiratory rate = 10 bpm, inspiratory-to-expiratory time ratio = 1:2, and fresh gas flow = 1 L/min air. N = 5 in each group. Control limits were determined from the first 150 minutes of battery power for each run and lower control limit = mean VT - 3SD. Battery depletion occurred when VT was below the lower control limit. Battery duration ranged from 185.8 (±3.2) minutes in the LC-1000 group to 233.3 (±3.6) minutes in the HC-750 group. Battery duration of the LC-1000 group was less than all others (p = 0.027). The differences among the non-LC-1000 groups were not clinically significant.
Department of Anesthesiology, Uniformed Services University, Room C-1090, 4301 Jones Bridge Road, Bethesda, MD 20814.
Dale F
DF
Szpisjak
Department of Anesthesiology, Fort Belvoir Community Hospital, 9300 DeWitt Loop, Fort Belvoir, VA 22060.
Anthony A
AA
Giberman
D000758:Anesthesia; D058868:Compressed Air; D011211:Electric Power Supplies; D006801:Humans; D008170:Lung Compliance; D012121:Respiration, Artificial / Q000295:instrumentation*; D013990:Tidal Volume; D013997:Time Factors; D012122:Ventilators, Mechanical