title: Instrumentation for analytical scale supercritical fluid chromatography.
abstract: Analytical scale supercritical fluid chromatography (SFC) is largely a sub-discipline of high performance liquid chromatography (HPLC), in that most of the hardware and software can be used for either technique. The aspects that separate the 2 techniques stem from the use of carbon dioxide (CO2) as the main component of the mobile phase in SFC. The high compressibility and low viscosity of CO2 mean that pumps, and autosamplers designed for HPLC either need to be modified or an alternate means of dealing with compressibility needs to be found. The inclusion of a back pressure regulator and a high pressure flow cell for any UV-Vis detector are also necessary. Details of the various approaches, problems and solutions are described. Characteristics, such as adiabatic vs. isothermal compressibility, thermal gradients, and refractive index issues are dealt with in detail.
authors.affiliation: SFC Solutions., Inc., USA. Electronic address: tabergersfc@aol.com.
authors.forename: Terry A
authors.initials: TA
authors.lastname: Berger
mesh_terms: D002245:Carbon Dioxide / Q000737:chemistry; D002851:Chromatography, High Pressure Liquid / Q000295:instrumentation / Q000379:methods; D025924:Chromatography, Supercritical Fluid / Q000295:instrumentation* / Q000379:methods; D011312:Pressure; D014783:Viscosity
chemical_list: D002245:Carbon Dioxide
keywords: Chillers; Compressibility; Extra-column band broadening; Modify autosampler; Pumps; Stagnant air oven