Ultrasound Velocity Measurement in a Liquid Metal Electrode.
A growing number of electrochemical technologies depend on fluid flow, and often that fluid is opaque. Measuring the flow of an opaque fluid is inherently more difficult than measuring the flow of a transparent fluid, since optical methods are not applicable. Ultrasound can be used to measure the velocity of an opaque fluid, not only at isolated points, but at hundreds or thousands of points arrayed along lines, with good temporal resolution. When applied to a liquid metal electrode, ultrasound velocimetry involves additional challenges: high temperature, chemical activity, and electrical conductivity. Here we describe the experimental apparatus and methods that overcome these challenges and allow the measurement of flow in a liquid metal electrode, as it conducts current, at operating temperature. Temperature is regulated within ±2 °C using a Proportional-Integral-Derivative (PID) controller that powers a custom-built furnace. Chemical activity is managed by choosing vessel materials carefully and enclosing the experimental setup in an argon-filled glovebox. Finally, unintended electrical paths are carefully prevented. An automated system logs control settings and experimental measurements, using hardware trigger signals to synchronize devices. This apparatus and these methods can produce measurements that are impossible with other techniques, and allow optimization and control of electrochemical technologies like liquid metal batteries.
Department of Mechanical Engineering, University of Rochester.
Adalberto
A
Perez
Department of Mechanical Engineering, University of Rochester; d.h.kelley@rochester.edu.
Douglas H
DH
Kelley
D004566:Electrodes; D019220:High-Energy Shock Waves; D008670:Metals / Q000737:chemistry*; D010084:Oxidation-Reduction; D012212:Rheology / Q000379:methods*; D013696:Temperature; D014465:Ultrasonics / Q000379:methods*
D008670:Metals