Improved Filtering of Pulse Oximeter Monitoring Alarms in the Neonatal ICU: Bedside Significance.
BACKGROUND
The OxiMax N-600x containing SatSeconds alarm management software was designed to assist clinicians in discriminating nuisance alarms from those that are clinically relevant. Instead of sounding an alarm the moment the oxygen saturation reading violates the upper or lower limit settings, a magnitude and duration of tolerance can be set. The primary objective was to study the proportion of nuisance alarms relative to the proportion of clinically relevant alarms being filtered under 4 different SatSeconds alarm settings (ie, 10, 25, 50, and 100) in the neonatal intensive care environment.


METHODS
This is an observational prospective study of 50 infants cared for in 3 large neonatal ICUs. Infants were monitored for 4 continuous h each by a study observer with the study monitor, in addition to standard multiparameter monitors. The performance of the SatSeconds alarm was compared with nursing intervention.


RESULTS
The area under the receiver operating characteristic curve (95% CI) for SatSeconds alarm settings when compared with nursing intervention were as follows: 0.61 (0.57-0.66) when the SatSeconds alarm setting was off, 0.63 (0.59-0.68) for the 10 SatSeconds alarm setting, 0.64 (0.59-0.69) for the 25 SatSeconds alarm setting, 0.64 (0.59-0.69) for the 50 SatSeconds alarm setting, and 0.63 (0.58-0.68) for the 100 SatSeconds alarm setting, respectively.


CONCLUSIONS
The SatSeconds feature of the OxiMax N-600x pulse oximeter reduced some nuisance alarms; however, its specificity to nurse-identified desaturation events does not significantly improve with lengthening SatSeconds alarm settings.
Department of Pediatrics, Vanderbilt University School of Medicine, Nashville, Tennessee. beatrice.stefanescu@Vanderbilt.edu.
Beatrice M
BM
Stefanescu
Department of Pediatrics, Wake Forest School of Medicine, Winston-Salem, North Carolina.
T Michael
TM
O'Shea
Clinical Affairs, Respiratory and Monitoring Solutions, Covidien, Boulder, Colorado.
Fran
F
Haury
Department of Pediatrics, University of Alabama at Birmingham, Birmingham, Alabama.
Waldemar A
WA
Carlo
Department of Biostatistics, Vanderbilt University School of Medicine, Nashville, Tennessee.
James C
JC
Slaughter
D019540:Area Under Curve; D056902:Clinical Alarms; D005189:False Positive Reactions; D005260:Female; D006801:Humans; D000860:Hypoxia / Q000175:diagnosis*; D007223:Infant; D007231:Infant, Newborn; D015931:Intensive Care, Neonatal; D008297:Male; D008991:Monitoring, Physiologic / Q000295:instrumentation*; D010092:Oximetry / Q000295:instrumentation*; D011446:Prospective Studies; D012372:ROC Curve; D012984:Software
Nellcor OxiMax N-600x; SatSeconds alarm settings; SpO2; nuisance alarms