Metal artifact reduction in patients with dental implants using multispectral three-dimensional data acquisition for hybrid PET/MRI.
BACKGROUND
Hybrid positron emission tomography/magnetic resonance imaging (PET/MRI) shows high potential for patients with oropharyngeal cancer. Dental implants can cause substantial artifacts in the oral cavity impairing diagnostic accuracy. Therefore, we evaluated new MRI sequences with multi-acquisition variable-resonance image combination (MAVRIC SL) in comparison to conventional high-bandwidth techniques and in a second step showed the effect of artifact size on MRI-based attenuation correction (AC) with a simulation study.


METHODS
Twenty-five patients with dental implants prospectively underwent a trimodality PET/CT/MRI examination after informed consent was obtained under the approval of the local ethics committee. A conventional 3D gradient-echo sequence (LAVA-Flex) commonly used for MRI-based AC of PET (acquisition time of 14 s), a T1w fast spin-echo sequence with high bandwidth (acquisition time of 3.2 min), as well as MAVRIC SL sequence without and with increased phase acceleration (MAVRIC, acquisition time of 6 min; MAVRIC-fast, acquisition time of 3.5 min) were applied. The absolute and relative reduction of the signal void artifact was calculated for each implant and tested for statistical significance using the Wilcoxon signed-rank test. The effect of artifact size on PET AC was simulated in one case with a large tumor in the oral cavity. The relative difference of the maximum standardized uptake value (SUVmax) in the tumor was calculated for increasing artifact sizes centered over the second molar.


RESULTS
The absolute reduction of signal void from LAVA-Flex sequences to the T1-weighted fast spin-echo (FSE) sequences was 416 mm(2) (range 4 to 2,010 mm(2)) to MAVRIC 481 mm(2) (range 12 to 2,288 mm(2)) and to MAVRIC-fast 486 mm(2) (range 39 to 2,209 mm(2)). The relative reduction in signal void was significantly improved for both MAVRIC and MAVRIC-fast compared to T1 FSE (-75%/-78% vs. -62%, p < 0.001 for both). The relative error for SUVmax was negligible for artifacts of 0.5-cm diameter (-0.1%), but substantial for artifacts of 5.2-cm diameter (-33%).


CONCLUSIONS
MAVRIC-fast could become useful for artifact reduction in PET/MR for patients with dental implants. This might improve diagnostic accuracy especially for patients with tumors in the oropharynx and substantially improve accuracy of PET quantification.
Department of Medical Radiology, Division of Nuclear Medicine, University Hospital Zurich, Ramistr. 100, CH-8091, Zurich, Switzerland. J.Ggunzinger@bluemail.ch.
Jeanne M
JM
Gunzinger
Global MR Applications and Workflow, GE Healthcare, CH-8048, Zurich, Switzerland. Gaspar.Delso@usz.ch.
Gaspar
G
Delso
Department of Medical Radiology, Institute of Diagnostic and Interventional Radiology, University Hospital Zurich, Ramistr. 100, CH-8091, Zurich, Switzerland. Andreas.Boss@usz.ch.
Andreas
A
Boss
Department of Medical Radiology, Division of Nuclear Medicine, University Hospital Zurich, Ramistr. 100, CH-8091, Zurich, Switzerland. Miguel.Porto@usz.ch.
Miguel
M
Porto
Department of Medical Radiology, Division of Nuclear Medicine, University Hospital Zurich, Ramistr. 100, CH-8091, Zurich, Switzerland. hdavison2@nhs.net.
Helen
H
Davison
Department of Medical Radiology, Division of Nuclear Medicine, University Hospital Zurich, Ramistr. 100, CH-8091, Zurich, Switzerland. Gustav.vonSchulthess@usz.ch.
Gustav K
GK
von Schulthess
Department of Medical Radiology, Division of Nuclear Medicine, University Hospital Zurich, Ramistr. 100, CH-8091, Zurich, Switzerland. Martin.Huellner@usz.ch.|Department of Medical Radiology, Institute of Neuroradiology, University Hospital Zurich, Ramistr. 100, CH-8091, Zurich, Switzerland. Martin.Huellner@usz.ch.
Martin
M
Huellner
Department of Medical Radiology, Division of Nuclear Medicine, University Hospital Zurich, Ramistr. 100, CH-8091, Zurich, Switzerland. Paul.Stolzmann@usz.ch.|Department of Medical Radiology, Institute of Neuroradiology, University Hospital Zurich, Ramistr. 100, CH-8091, Zurich, Switzerland. Paul.Stolzmann@usz.ch.
Paul
P
Stolzmann
Department of Medical Radiology, Division of Nuclear Medicine, University Hospital Zurich, Ramistr. 100, CH-8091, Zurich, Switzerland. Patrick.Veit-Haibach@usz.ch.|Department of Medical Radiology, Institute of Diagnostic and Interventional Radiology, University Hospital Zurich, Ramistr. 100, CH-8091, Zurich, Switzerland. Patrick.Veit-Haibach@usz.ch.
Patrick
P
Veit-Haibach
Department of Medical Radiology, Division of Nuclear Medicine, University Hospital Zurich, Ramistr. 100, CH-8091, Zurich, Switzerland. irene.burger@usz.ch.|Department of Medical Radiology, Institute of Diagnostic and Interventional Radiology, University Hospital Zurich, Ramistr. 100, CH-8091, Zurich, Switzerland. irene.burger@usz.ch.
Irene A
IA
Burger
Attenuation correction; Image noise; MAVRIC; Signal voids