title: A Side by Side Comparison of Bruker Biotyper and VITEK MS: Utility of MALDI-TOF MS Technology for Microorganism Identification in a Public Health Reference Laboratory.
abstract: Matrix-assisted laser desorption ionization-time of flight mass spectrometry (MALDI-TOF MS) has emerged as a rapid, highly accurate, and cost-effective method for routine identification of a wide range of microorganisms. We carried out a side by side comparative evaluation of the performance of Bruker Biotyper versus VITEK MS for identification of a large and diverse collection of microorganisms. Most difficult and/or unusual microorganisms, as well as commonly encountered microorganisms were selected, including Gram-positive and negative bacteria, mycobacteria, actinomycetes, yeasts and filamentous fungi. Six hundred forty two strains representing 159 genera and 441 species from clinical specimens previously identified at the Laboratoire de santé publique du Québec (LSPQ) by reference methods were retrospectively chosen for the study. They included 254 Gram-positive bacteria, 167 Gram-negative bacteria, 109 mycobacteria and aerobic actinomycetes and 112 yeasts and moulds. MALDI-TOF MS analyses were performed on both systems according to the manufacturer's instructions. Of the 642 strains tested, the name of the genus and / or species of 572 strains were referenced in the Bruker database while 406 were present in the VITEK MS IVD database. The Biotyper correctly identified 494 (86.4%) of the strains, while the VITEK MS correctly identified 362 (92.3%) of the strains (excluding 14 mycobacteria that were not tested). Of the 70 strains not present in the Bruker database at the species level, the Biotyper correctly identified 10 (14.3%) to the genus level and 2 (2.9%) to the complex/group level. For 52 (74.2%) strains, we obtained no identification, and an incorrect identification was given for 6 (8.6%) strains. Of the 178 strains not present in the VITEK MS IVD database at the species level (excluding 71 untested mycobacteria and actinomycetes), the VITEK MS correctly identified 12 (6.8%) of the strains each to the genus and to the complex/group level. For 97 (54.5%) strains, no identification was given and for 69 (38.7%) strains, an incorrect identification was obtained. Our study demonstrates that both systems gave a high level (above 85%) of correct identification for a wide range of microorganisms. However, VITEK MS gave more misidentification when the microorganism analysed was not present in the database, compared to Bruker Biotyper. This should be taken into account when this technology is used alone for microorganism identification in a public health laboratory, where isolates received are often difficult to identify and/or unusual microorganisms.
authors.affiliation: Laboratoire de santé publique du Québec / Institut national de santé publique du Québec, Sainte-Anne-de-Bellevue, Québec, Canada.
authors.forename: Simon
authors.initials: S
authors.lastname: Lévesque
authors.affiliation: Laboratoire de santé publique du Québec / Institut national de santé publique du Québec, Sainte-Anne-de-Bellevue, Québec, Canada.
authors.forename: Philippe J
authors.initials: PJ
authors.lastname: Dufresne
authors.affiliation: Laboratoire de santé publique du Québec / Institut national de santé publique du Québec, Sainte-Anne-de-Bellevue, Québec, Canada.
authors.forename: Hafid
authors.initials: H
authors.lastname: Soualhine
authors.affiliation: Laboratoire de santé publique du Québec / Institut national de santé publique du Québec, Sainte-Anne-de-Bellevue, Québec, Canada.
authors.forename: Marc-Christian
authors.initials: MC
authors.lastname: Domingo
authors.affiliation: Laboratoire de santé publique du Québec / Institut national de santé publique du Québec, Sainte-Anne-de-Bellevue, Québec, Canada.|Département de Microbiologie, Immunologie et Infectiologie, Université de Montréal, Montréal, Québec, Canada.
authors.forename: Sadjia
authors.initials: S
authors.lastname: Bekal
authors.affiliation: Laboratoire de santé publique du Québec / Institut national de santé publique du Québec, Sainte-Anne-de-Bellevue, Québec, Canada.
authors.forename: Brigitte
authors.initials: B
authors.lastname: Lefebvre
authors.affiliation: Laboratoire de santé publique du Québec / Institut national de santé publique du Québec, Sainte-Anne-de-Bellevue, Québec, Canada.|Centre de Recherche du Centre Hospitalier de l'Université de Montréal, Montréal, Québec, Canada.
authors.forename: Cécile
authors.initials: C
authors.lastname: Tremblay
mesh_terms: D039903:Actinobacteria / Q000145:classification / Q000302:isolation & purification; D015373:Bacterial Typing Techniques / Q000379:methods; D064790:Clinical Laboratory Services; D019411:Clinical Laboratory Techniques / Q000379:methods*; D005658:Fungi / Q000145:classification / Q000302:isolation & purification; D006090:Gram-Negative Bacteria / Q000145:classification / Q000302:isolation & purification; D006801:Humans; D009161:Mycobacterium / Q000145:classification / Q000302:isolation & purification; D016533:Mycological Typing Techniques / Q000379:methods; D011634:Public Health; D011792:Quebec; D015203:Reproducibility of Results; D019032:Spectrometry, Mass, Matrix-Assisted Laser Desorption-Ionization / Q000379:methods*; D015003:Yeasts / Q000145:classification / Q000302:isolation & purification