title: Dosimetric effects of positioning shifts using 6D-frameless stereotactic Brainlab system in hypofractionated intracranial radiotherapy.
abstract: Dosimetric consequences of positional shifts were studied using frameless Brainlab ExacTrac X-ray system for hypofractionated (3 or 5 fractions) intracranial stereo-tactic radiotherapy (SRT). SRT treatments of 17 patients with metastatic intracranial tumors using the stereotactic system were retrospectively investigated. The treatments were simulated in a treatment planning system by modifying planning parameters with a matrix conversion technique based on positional shifts for initial infrared (IR)-based setup (XC: X-ray correction) and post-correction (XV: X-ray verification). The simulation was implemented with (a) 3D translational shifts only and (b) 6D translational and rotational shifts for dosimetric effects of angular correction. Mean translations and rotations (± 1 SD) of 77 fractions based on the initial IR setup (XC) were 0.51 ± 0.86 mm (lateral), 0.30 ± 1.55 mm (longitudinal), and -1.63 ± 1.00 mm (vertical); -0.53° ± 0.56° (pitch), 0.42° ± 0.60° (roll), and 0.44°± 0.90° (yaw), respectively. These were -0.07 ± 0.24 mm, -0.07 ± 0.25 mm, 0.06± 0.21 mm, 0.04° ± 0.23°, 0.00° ± 0.30°, and -0.02° ± 0.22°, respectively, for the postcorrection (XV). Substantial degradation of the treatment plans was observed in D95 of PTV (2.6% ± 3.3%; simulated treatment versus treatment planning), Dmin of PTV (13.4% ± 11.6%), and Dmin of CTV (2.8% ± 3.8%, with the maximum error of 10.0%) from XC, while dosimetrically negligible changes (< 0.1%) were detected for both CTV and PTV from XV simulation. 3D angular correction significantly improved CTV dose coverage when the total angular shifts (|pitch| + |roll| + |yaw|) were greater than 2°. With the 6D stereoscopic X-ray verification imaging and frameless immobilization, submillimeter and subdegree accuracy is achieved with negligible dosimetric deviations. 3D angular correction is required when the angular deviation is substantial. A CTV-to-PTV safety margin of 2 mm is large enough to prevent deterioration of CTV coverage.
authors.affiliation: University of Oklahoma Health Sciences Center. hosang.jin@gmail.com.
authors.forename: Hosang
authors.initials: H
authors.lastname: Jin
authors.forename: Vance P
authors.initials: VP
authors.lastname: Keeling
authors.forename: Imad
authors.initials: I
authors.lastname: Ali
authors.forename: Salahuddin
authors.initials: S
authors.lastname: Ahmad
mesh_terms: D001932:Brain Neoplasms / Q000601:surgery*; D006801:Humans; D021621:Imaging, Three-Dimensional / Q000379:methods; D056888:Patient Positioning; D011874:Radiometry / Q000379:methods; D016634:Radiosurgery; D011879:Radiotherapy Dosage; D011880:Radiotherapy Planning, Computer-Assisted / Q000379:methods*; D061106:Radiotherapy Setup Errors / Q000517:prevention & control*; D050397:Radiotherapy, Intensity-Modulated / Q000379:methods*; D012189:Retrospective Studies