title: Hadamard-transform fluorescence-lifetime imaging.
abstract: We discuss a Hadamard-transform-based fluorescence-lifetime-imaging (HT-FLI) technique for fluorescence-lifetime-imaging microscopy (FLIM). The HT-FLI uses a Fourier-transform phase-modulation fluorometer (FT-PMF) for fluorescence-lifetime measurements, where the modulation frequency of the excitation light is swept linearly in frequency from zero to a specific maximum during a fixed duration of time. Thereafter, fluorescence lifetimes are derived through Fourier transforms for the fluorescence and reference waveforms. The FT-PMF enables the analysis of multi-component samples simultaneously. HT imaging uses electronic exchange of HT illumination mask patterns, and a high-speed, high-sensitivity photomultiplier, to eliminate frame-rate issues that accompany two-dimensional image detectors.
authors.forename: Takahiko
authors.initials: T
authors.lastname: Mizuno
authors.forename: Tetsuo
authors.initials: T
authors.lastname: Iwata