title: TRIO Platform: A Novel Low Profile In vivo Imaging Support and Restraint System for Mice.
abstract: High resolution, in vivo optical imaging of the mouse brain over time often requires anesthesia, which necessitates maintaining the animal's body temperature and level of anesthesia, as well as securing the head in an optimal, stable position. Controlling each parameter usually requires using multiple systems. Assembling multiple components into the small space on a standard microscope stage can be difficult and some commercially available parts simply do not fit. Furthermore, it is time-consuming to position an animal in the identical position over multiple imaging sessions for longitudinal studies. This is especially true when using an implanted gradient index (GRIN) lens for deep brain imaging. The multiphoton laser beam must be parallel with the shaft of the lens because even a slight tilt of the lens can degrade image quality. In response to these challenges, we have designed a compact, integrated in vivo imaging support system to overcome the problems created by using separate systems during optical imaging in mice. It is a single platform that provides (1) sturdy head fixation, (2) an integrated gas anesthesia mask, and (3) safe warm water heating. This THREE-IN-ONE (TRIO) Platform has a small footprint and a low profile that positions a mouse's head only 20 mm above the microscope stage. This height is about one half to one third the height of most commercially available immobilization devices. We have successfully employed this system, using isoflurane in over 40 imaging sessions with an average of 2 h per session with no leaks or other malfunctions. Due to its smaller size, the TRIO Platform can be used with a wider range of upright microscopes and stages. Most of the components were designed in SOLIDWORKS® and fabricated using a 3D printer. This additive manufacturing approach also readily permits size modifications for creating systems for other small animals.
authors.affiliation: Integrated Neuroscience and Imaging Lab, Center for Biomedical Engineering and Rehabilitation Sciences, Louisiana Tech UniversityRuston, LA, USA; School of Biological and Health Systems Engineering, Ira A. Fulton School of Engineering, Arizona State UniversityTempe, AZ, USA.
authors.forename: Vladislav
authors.initials: V
authors.lastname: Voziyanov
authors.affiliation: Integrated Neuroscience and Imaging Lab, Center for Biomedical Engineering and Rehabilitation Sciences, Louisiana Tech University Ruston, LA, USA.
authors.forename: Benjamin S
authors.initials: BS
authors.lastname: Kemp
authors.affiliation: Integrated Neuroscience and Imaging Lab, Center for Biomedical Engineering and Rehabilitation Sciences, Louisiana Tech University Ruston, LA, USA.
authors.forename: Chelsea A
authors.initials: CA
authors.lastname: Dressel
authors.affiliation: Integrated Neuroscience and Imaging Lab, Center for Biomedical Engineering and Rehabilitation Sciences, Louisiana Tech University Ruston, LA, USA.
authors.forename: Kayla
authors.initials: K
authors.lastname: Ponder
authors.affiliation: Integrated Neuroscience and Imaging Lab, Center for Biomedical Engineering and Rehabilitation Sciences, Louisiana Tech University Ruston, LA, USA.
authors.forename: Teresa A
authors.initials: TA
authors.lastname: Murray
keywords: anesthesia; astrocytes; cranial window; gradient index lens; head restraint; in vivo imaging; multiphoton microscopy; warming pad