title: Monte Carlo calculations and experimental measurements of the TG-43U1-recommended dosimetric parameters of 125I (Model IR-Seed2) brachytherapy source.
abstract: A new design of 125I (Model IR-Seed2) brachytherapy source has been manufactured recently at the Applied Radiation Research School, Nuclear Science and Technology Research Institute in Iran. The source consists of six resin beads (0.5 mm diameter) that are sealed in a cylindrical titanium capsule of 0.7 mm internal and 0.8 mm external diameters. This work aims to evaluate the dosimetric parameters of the newly designed 125I source using experimental measurements and Monte Carlo (MC) simulations. Dosimetric characteristics (dose rate constant, radial dose function, and 2D and 1D anisotropy functions) of the IR-Seed2 were determined using experimental measurements and MC simulations following the recommendations by the Task Group 43 (TG-43U1) report of the American Association of Physicists in Medicine (AAPM). MC simulations were performed using the MCNP5 code in water and Plexiglas, and experimental measurements were carried out using thermoluminescent dosimeters (TLD-GR207A) in Plexiglas phantoms. The measured dose to water in Plexiglas data were used for verification of the accuracy of the source and phantom geometry in the Monte Carlo simulations. The final MC simulated data to water in water were recommended for clinical applications. The MC calculated dose rate constant (Λ) of the IR-Seed2 125I seed in water was found to be 0.992 ± 0.025 cGy h-1U-1. Additionally, its radial dose function by line and point source approximations, gL(r) and gp(r), calculated for distances from 0.1 cm to 7 cm. The values of gL(r) at radial distances from 0.5 cm to 5 cm were measured in a Plexiglas phantom to be between 1.212 and 0.413. The calculated and measured of values for 2D anisotropy function, F(r, θ), were obtained for the radial distances ranging from 1.5 cm to 5 cm and angular range of 0°-90° in a Plexiglas phantom. Also, the 2D anisotropy function was calculated in water for the clinical application. The results of these investigations show that the uncertainty of the experimental data is within ± 7% between the measured and simulated data in Plexiglas. Based on these results, the MC-simulated dosimetric parameters of the new 125I source model in water are presented for its clinical applications in brachytherapy treatments.
authors.affiliation: Islamic Azad University. s.sheikholeslami67@yahoo.com.
authors.forename: Sahar
authors.initials: S
authors.lastname: Sheikholeslami
authors.forename: Hasan Ali
authors.initials: HA
authors.lastname: Nedaie
authors.forename: Mahdi
authors.initials: M
authors.lastname: Sadeghi
authors.forename: Hosein
authors.initials: H
authors.lastname: Pourbeigy
authors.forename: Sohrab
authors.initials: S
authors.lastname: Shahzadi
authors.forename: Mehdi
authors.initials: M
authors.lastname: Zehtabian
authors.forename: Mohsen
authors.initials: M
authors.lastname: Hasani
authors.forename: Ali S
authors.initials: AS
authors.lastname: Meigooni
mesh_terms: D016880:Anisotropy; D001918:Brachytherapy / Q000295:instrumentation* / Q000592:standards*; D006801:Humans; D007457:Iodine Radioisotopes; D009010:Monte Carlo Method; D010165:Palladium; D019047:Phantoms, Imaging; D011879:Radiotherapy Dosage; D013819:Thermoluminescent Dosimetry / Q000592:standards*
chemical_list: D007457:Iodine Radioisotopes; D010165:Palladium