title: [Evaluating the accuracy of three-dimensional reconstruction of the intercuspal position for dentition casts aided by a mechanical appliance].
abstract: OBJECTIVE
  To develop a aided mechanical appliance for rapid reconstruction of three-dimensional(3D)relationship of dentition model after scanning and evaluation of its accuracy.
  
  
  METHODS
  The appliance was designed by forward engineering software and fabricated by a high precision computer numerical control(CNC)system. It contained upper and lower body, magnetic pedestal and three pillars. Nine 3 mm diameter hemispheres were distributed equally on the axial surface of each pedestal. Faro Edge 1.8m was used to directly obtain center of each hemisphere(contact method), defined as known center. A pair of die-stone standard dentition model were fixed in intercuspal position and then fixed on the magnetic pedestals with low expansion ratio plaster. Activity 880 dental scanner was used to scan casts after the plaster was completely set. In Geomagic 2012, the centers of each hemisphere were fitted and defined as scanning centers. Scanning centers were aligned to known centers by reference point system to finish the 3D reconstruction of the intercuspal occlusion for the dentition casts. An observation coordinate system was interactively established. The straight-line distances in the X(coronal), Y(saggital), and Z(vertical)between the remaining 6 pairs of center points derived from contact method and fitting method were measured respectively and analyzed using a paired t-test.
  
  
  RESULTS
  The differences of the straight-line distances of the remaining 6 pairs of center points between the two methods were X:(-0.05±0.10)mm, Y:(0.02±0.06)mm, and Z:(0.01 ± 0.05)mm. The results of paired t-test showed no significant differences(P>0.05).
  
  
  CONCLUSIONS
  The mechanical appliance can help to reconstruct 3D jaw relation by scanning single upper and lower dentition model with usual commercial available dental cast scanning system.
authors: Center of Digital Dentistry, Faculty of Prosthodontics, Peking University School and Hospital of Stomatology& National Engineering Laboratory for Digital and Material Technology of Stomatology & Research Center of Engineering and Technology for Digital Dentistry of Ministry of Health & Beijing Key Laboratory of Digital Stomatology, Beijing 100081, China.
authors: Z W
authors: ZW
authors: Hu
authors: Center of Digital Dentistry, Faculty of Prosthodontics, Peking University School and Hospital of Stomatology& National Engineering Laboratory for Digital and Material Technology of Stomatology & Research Center of Engineering and Technology for Digital Dentistry of Ministry of Health & Beijing Key Laboratory of Digital Stomatology, Beijing 100081, China.
authors: W W
authors: WW
authors: Li
authors: Center of Digital Dentistry, Faculty of Prosthodontics, Peking University School and Hospital of Stomatology& National Engineering Laboratory for Digital and Material Technology of Stomatology & Research Center of Engineering and Technology for Digital Dentistry of Ministry of Health & Beijing Key Laboratory of Digital Stomatology, Beijing 100081, China.
authors: X Y
authors: XY
authors: Zhang
authors: Department of Medical Equipment Management, Peking University School and Hospital of Stomatology, National Engineering Laboratory for Digital and Material Technology of Stomatology & Beijing Key Laboratory of Digital Stomatology, Beijing 100081, China.
authors: B L
authors: BL
authors: Fan
authors: Center of Digital Dentistry, Faculty of Prosthodontics, Peking University School and Hospital of Stomatology& National Engineering Laboratory for Digital and Material Technology of Stomatology & Research Center of Engineering and Technology for Digital Dentistry of Ministry of Health & Beijing Key Laboratory of Digital Stomatology, Beijing 100081, China.
authors: Y
authors: Y
authors: Wang
authors: Center of Digital Dentistry, Faculty of Prosthodontics, Peking University School and Hospital of Stomatology& National Engineering Laboratory for Digital and Material Technology of Stomatology & Research Center of Engineering and Technology for Digital Dentistry of Ministry of Health & Beijing Key Laboratory of Digital Stomatology, Beijing 100081, China.
authors: Y C
authors: YC
authors: Sun
mesh_terms: D003766:Dental Occlusion; D003817:Dentition; D006801:Humans; D007091:Image Processing, Computer-Assisted; D003765:Models, Dental; D012984:Software