Towards a Reduced-Wire Interface for CMUT-Based Intravascular Ultrasound Imaging Systems.
Having intravascular ultrasound (IVUS) imaging capability on guide wires used in cardiovascular interventions may eliminate the need for separate IVUS catheters and expand the use of IVUS in a larger portion of the vasculature. High frequency capacitive micro machined ultrasonic transducer (CMUT) arrays should be integrated with interface electronics and placed on the guide wire for this purpose. Besides small size, this system-on-a-chip (SoC) front-end should connect to the back-end imaging system with a minimum number of wires to preserve the critical mechanical properties of the guide wire. We present a 40 MHz CMUT array interface SoC, which will eventually use only two wires for power delivery and transmits image data using a combination of analog-to-time conversion (ATC) and an impulse radio ultra-wideband (IR-UWB) wireless link. The proof-of-concept prototype ASIC consumes only 52.8 mW and occupies 4.07 [Formula: see text] in a 0.35- [Formula: see text] standard CMOS process. A rectifier and regulator power the rest of the SoC at 3.3 V from a 10 MHz power carrier that is supplied through a 2.4 m micro-coax cable with an overall efficiency of 49.1%. Echo signals from an 8-element CMUT array are amplified by a transimpedance amplifier (TIA) array and down-converted to baseband by quadrature sampling using a 40 MHz clock, derived from the power carrier. The ATC generates pulse-width-modulated (PWM) samples at 2 × 10 MS/s with 6 bit resolution, while the entire system achieved 5.1 ENOB. Preliminary images from the prototype system are presented, and alternative data transmission and possible future directions towards practical implementation are discussed.
Jaemyung
J
Lim
Coskun
C
Tekes
F Levent
FL
Degertekin
Maysam
M
Ghovanloo
D000669:Amplifiers, Electronic; D001808:Blood Vessels / Q000000981:diagnostic imaging*; D004867:Equipment Design; D006801:Humans; D014159:Transducers; D018084:Ultrasonography, Interventional / Q000295:instrumentation*
Int J Cardiovasc Imaging. 2011 Feb;27(2):215-24
21327914
IEEE Trans Ultrason Ferroelectr Freq Control. 2007 Sep;54(9):1792-801
17941385
Conf Proc IEEE Eng Med Biol Soc. 2014;2014:5080-4
25571135
IEEE Trans Ultrason Ferroelectr Freq Control. 2014 Feb;61(2):239-50
24474131
Circ Cardiovasc Interv. 2012 Apr;5(2):312-7
22511740
Conf Proc IEEE Eng Med Biol Soc. 2014;2014:3134-7
25570655
J Am Coll Cardiol. 2012 Dec 4;60(22):2337-9
23194945
IEEE Trans Neural Syst Rehabil Eng. 2009 Aug;17(4):354-63
19497823
IEEE Trans Ultrason Ferroelectr Freq Control. 2013 Nov;60(11):2426-39
24158297
Ann Thorac Surg. 2008 Oct;86(4):1391-8
18805213
IEEE J Biomed Health Inform. 2015 May;19(3):938-48
25861089
IEEE Trans Ultrason Ferroelectr Freq Control. 2008 Dec;55(12):2651-60
19126489
Eur Heart J. 2010 Oct;31(20):2456-69
20823109
IEEE Trans Ultrason Ferroelectr Freq Control. 1992;39(3):366-80
18267647
Phys Med Biol. 1996 Nov;41(11):2251-69
8938025