A soft decoding algorithm and hardware implementation for the visual prosthesis based on high order soft demodulation.
BACKGROUND
High order modulation and demodulation technology can solve the frequency requirement between the wireless energy transmission and data communication. In order to achieve reliable wireless data communication based on high order modulation technology for visual prosthesis, this work proposed a Reed-Solomon (RS) error correcting code (ECC) circuit on the basis of differential amplitude and phase shift keying (DAPSK) soft demodulation. Firstly, recognizing the weakness of the traditional DAPSK soft demodulation algorithm based on division that is complex for hardware implementation, an improved phase soft demodulation algorithm for visual prosthesis to reduce the hardware complexity is put forward. Based on this new algorithm, an improved RS soft decoding method is hence proposed. In this new decoding method, the combination of Chase algorithm and hard decoding algorithms is used to achieve soft decoding. In order to meet the requirements of implantable visual prosthesis, the method to calculate reliability of symbol-level based on multiplication of bit reliability is derived, which reduces the testing vectors number of Chase algorithm. The proposed algorithms are verified by MATLAB simulation and FPGA experimental results. During MATLAB simulation, the biological channel attenuation property model is added into the ECC circuit.


RESULTS
The data rate is 8 Mbps in the MATLAB simulation and FPGA experiments. MATLAB simulation results show that the improved phase soft demodulation algorithm proposed in this paper saves hardware resources without losing bit error rate (BER) performance. Compared with the traditional demodulation circuit, the coding gain of the ECC circuit has been improved by about 3 dB under the same BER of [Formula: see text]. The FPGA experimental results show that under the condition of data demodulation error with wireless coils 3 cm away, the system can correct it. The greater the distance, the higher the BER. Then we use a bit error rate analyzer to measure BER of the demodulation circuit and the RS ECC circuit with different distance of two coils. And the experimental results show that the RS ECC circuit has about an order of magnitude lower BER than the demodulation circuit when under the same coils distance. Therefore, the RS ECC circuit has more higher reliability of the communication in the system.


CONCLUSIONS
The improved phase soft demodulation algorithm and soft decoding algorithm proposed in this paper enables data communication that is more reliable than other demodulation system, which also provide a significant reference for further study to the visual prosthesis system.
Department of Electronic Engineering, Xi'an University of Technology, Jinhua Road No. 5, Xi'an, 710048, China. yangyuan@xaut.edu.cn.
Yuan
http://orcid.org/0000-0002-4225-5226
Y
Yang
Department of Electronic Engineering, Xi'an University of Technology, Jinhua Road No. 5, Xi'an, 710048, China.
Nannan
N
Quan
Department of Electronic Engineering, Xi'an University of Technology, Jinhua Road No. 5, Xi'an, 710048, China.
Jingjing
J
Bu
Department of Electronic Engineering, Xi'an University of Technology, Jinhua Road No. 5, Xi'an, 710048, China.
Xueping
X
Li
Department of Electronic Engineering, Xi'an University of Technology, Jinhua Road No. 5, Xi'an, 710048, China.
Ningmei
N
Yu
Bit error rate; Differential amplitude phase-shift keying; Reed–Solomon code; Visual prosthesis
Conf Proc IEEE Eng Med Biol Soc. 2009;2009:4840-3
19963861
Brain Res. 2016 Jan 1;1630:208-24
26348986
IEEE J Biomed Health Inform. 2014 Nov;18(6):1848-56
25375682
IEEE Trans Biomed Circuits Syst. 2015 Feb;9(1):1-11
24760945
Brain Res. 2015 Jan 21;1595:51-73
25446438
Conf Proc IEEE Eng Med Biol Soc. 2011;2011:2882-5
22254943
Conf Proc IEEE Eng Med Biol Soc. 2011;2011:4010-3
22255219
Conf Proc IEEE Eng Med Biol Soc. 2011;2011:8356-60
22256285