Extension of the NIST AC-DC Difference Calibration Service for Current to 100 kHz.
The NIST calibration service for ac-dc difference of thermal current converters relies on multijunction thermal converters as the primary standards, and various thermal converters and thermoelements (TEs) as the reference and working standards. Calibrations are performed by comparing the ac-dc difference of a customer's thermal current converter to the ac-dc difference of a NIST standard current converter. Typical artifacts accepted for calibration include single-junction thermoelements, multijunction thermal converters, and transfer shunts for use with TEs. This paper describes the standards on which the calibration service is based and the results of the study to characterize the NIST standards over the extended frequency range from 50 kHz to 100 kHz at currents from 1 mA to 20 A. The general method for the frequency extension at high frequency involves the use of thermoelements in the 5 mA range, with small frequency dependence, as the starting point for build-up and build-down chains to cover the whole range from 1 mA to 20 A.
National Institute of Standards and Technology, Gaithersburg, MD 20899-0001.
Joseph R
JR
Kinard
National Institute of Standards and Technology, Gaithersburg, MD 20899-0001.
Thomas E
TE
Lipe
National Institute of Standards and Technology, Gaithersburg, MD 20899-0001.
Clifton B
CB
Childers
ac current; ac-dc difference; current calibrations; current shunts; thermal converter; thermal current converter; thermoelement