ITER-like antenna capacitors voltage probes: Circuit/electromagnetic calculations and calibrations.
The analyses illustrated in this manuscript have been performed in order to provide the required data for the amplitude-and-phase calibration of the D-dot voltage probes used in the ITER-like antenna at the Joint European Torus tokamak. Their equivalent electrical circuit has been extracted and analyzed, and it has been compared to the one of voltage probes installed in simple transmission lines. A radio-frequency calibration technique has been formulated and exact mathematical relations have been derived. This technique mixes in an elegant fashion data extracted from measurements and numerical calculations to retrieve the calibration factors. The latter have been compared to previous calibration data with excellent agreement proving the robustness of the proposed radio-frequency calibration technique. In particular, it has been stressed that it is crucial to take into account environmental parasitic effects. A low-frequency calibration technique has been in addition formulated and analyzed in depth. The equivalence between the radio-frequency and low-frequency techniques has been rigorously demonstrated. The radio-frequency calibration technique is preferable in the case of the ITER-like antenna due to uncertainties on the characteristics of the cables connected at the inputs of the voltage probes. A method to extract the effect of a mismatched data acquisition system has been derived for both calibration techniques. Finally it has been outlined that in the case of the ITER-like antenna voltage probes can be in addition used to monitor the currents at the inputs of the antenna.
CEA, IRFM, F-13108 St-Paul-Lez-Durance, France.
W
W
Helou
LPP-ERM/KMS, TEC partner, Brussels, Belgium.
P
P
Dumortier
LPP-ERM/KMS, TEC partner, Brussels, Belgium.
F
F
Durodié
CEA, IRFM, F-13108 St-Paul-Lez-Durance, France.
G
G
Lombard
CCFE, Culham Science Centre, Abingdon OX14 3DB, United Kingdom.
K
K
Nicholls
EUROfusion Consortium, JET, Culham Science Centre, Abingdon OX14 3DB, United KingdomCEA, IRFM, F-13108 St-Paul-Lez-Durance, FranceLPP-ERM/KMS, TEC partner, Brussels, BelgiumCCFE, Culham Science Centre, Abingdon OX14 3DB, United Kingdom.