title: High Power CW Wattmeter Calibration at NIST.
abstract: The National Institute of Standards and Technology has established a measurement capability to support high power systems and devices. The automated wattmeter calibration system operates at power levels of 1 to 1000 W for frequencies from 1 to 30 MHz and 1 to 500 W from 30 to 400 MHz. A cascaded coupler technique is used to extend power measurements to high levels which are traccable to a 10 mW standard thermistor mount. This technique uses an arrangement of nominal 10, 20, 30, 40, and 50 dB couplers with sidearm power meters. The initial step transfers the calibration of the 10 mW standard to the 10 dB coupler/power meter. The standard is then replaced with a wattmeter to be calibrated. RF power is increased 10 dB and the calibration is transferred to the adjacent 20 dB eoupler/power meter. This sequence is repeated with the remaining coupler/power meters until the wattmeter is calibrated at the desired power levels and frequencies. Power ratios calculated from simultaneous power measurements made at each transfer are used to calculate the incident power at the wattmeter. Due to nonideal components, corrections are made for nonlinearities, mismatch, and other errors. Two types of wattmeters have been evaluated at selected frequencies and power levels. Total uncertainties are based on the random and systematic components.
authors.affiliation: National Institute of Standards and Technology, Boulder, CO 80303.
authors.forename: Gregorio
authors.initials: G
authors.lastname: Rebuldela
authors.affiliation: National Institute of Standards and Technology, Boulder, CO 80303.
authors.forename: Jeffrey A
authors.initials: JA
authors.lastname: Jargon
keywords: automated; calibration; cascaded; continuous wave; coupler; high power; measurement; transfer; uncertainty; wattmeter