A microfabricated optically-pumped magnetic gradiometer.
We report on the development of a microfabricated atomic magnetic gradiometer based on optical spectroscopy of alkali atoms in the vapor phase. The gradiometer, which operates in the spin-exchange relaxation free regime, has a length of 60 mm and cross sectional diameter of 12 mm, and consists of two chip-scale atomic magnetometers which are interrogated by a common laser light. The sensor can measure differences in magnetic fields, over a 20 mm baseline, of 10 fT/[Formula: see text] at frequencies above 20 Hz. The maximum rejection of magnetic field noise is 1000 at 10 Hz. By use of a set of compensation coils wrapped around the sensor, we also measure the sensor sensitivity at several external bias field strengths up to 150 mG. This device is useful for applications that require both sensitive gradient field information and high common-mode noise cancellation.
D
D
Sheng
A R
AR
Perry
S P
SP
Krzyzewski
Time and Frequency Division, National Institute of Standards and Technology , 325 Broadway, Boulder, Colorado 80305, USA.
S
S
Geller
Time and Frequency Division, National Institute of Standards and Technology , 325 Broadway, Boulder, Colorado 80305, USA.
J
J
Kitching
S
S
Knappe