Quantitative x-ray microanalysis of model biological samples in the SEM using remote standards and the XPP analytical model.
It is shown that accurate x-ray microanalysis of frozen-hydrated and dry organic compounds, such as model biological samples, is possible with a silicon drift detector in combination with XPP (exponential model of Pouchou and Pichoir matrix correction) software using 'remote standards'. This type of analysis is also referred to as 'standardless analysis'. Analyses from selected areas or elemental images (maps) were identical. Improvements in x-ray microanalytical hardware and software, together with developments in cryotechniques, have made the quantitative analysis of cryoplaned frozen-hydrated biological samples in the scanning electron microscope a much simpler procedure. The increased effectiveness of pulse pile-up rejection renders the analysis of Na, with ultrathin window detectors, in the presence of very high concentrations of O, from ice, more accurate. The accurate analysis of Ca (2 mmol kg-1) in the presence of high concentrations of K is possible. Careful sublimation of surface frost from frozen-hydrated samples resulted in a small increase in analysed elemental concentrations. A more prolonged sublimation from the same resurfaced sample and other similar samples resulted in higher element concentrations.
Analytical Electron Microscopy Laboratory, Department of Ecology,Environment and Evolution, La Trobe University, Melbourne, Victoria, 3086, Australia.
Alan T
AT
Marshall
D001688:Biological Products / Q000737:chemistry*; D004577:Electron Probe Microanalysis / Q000379:methods* / Q000592:standards*; D004602:Elements; D007091:Image Processing, Computer-Assisted / Q000379:methods* / Q000592:standards*; D008855:Microscopy, Electron, Scanning / Q000379:methods* / Q000592:standards*
D001688:Biological Products; D004602:Elements
Biological/organic samples; X-ray microanalysis; XPP model; frozen-hydrated; silicon drift detector; standardless analysis