Biomechanical cadaveric comparison of patellar ligament suture protected by a steel cable versus a synthetic cable.
BACKGROUND
Purpose and hypothesis: Patellar ligament rupture is a rare disabling pathology requiring a surgical ligament suture protected by a frame. The gold standard is the steel cable, but its rigidity and the necessity of a surgical re-intervention for its removal render it unsatisfactory. The objective of this paper is to quantify the mechanical protection provided by the terylene® in comparison with steel.


METHODS
Twenty-four knees of 12 fresh frozen cadaveric subjects were divided into 2 homogeneous groups (terylene and steel) of 12 knees (mean age = 69.3 years). Proximal ligament repair was performed according to a three-tunnel transosseous reinsertion technique. Mechanical tests were performed in flexion to simulate movement of the knee. The interligament gap and the amplitude angulation of the knee were measured by a system of extensometer and optical goniometer. Mechanical analysis permitted calculation of flexion amplitude for a ligament gap of 1 and 2 mm taking as initial angle the adjusting angle of pretension of the protection frame. Study of deformations of frames was performed. Statistical analysis was performed with a Wilcoxon Mann Whitney test.


RESULTS
There is no significant difference in protection of the ligament suture between the "terylene" and "steel" groups. Mean flexion amplitudes (mΔF) show no significant differences between the 2 groups for a distension of the suture of 1 mm (m ΔF terylene1 = 4.74 °; mΔF steel1 = 5.91°; p = 0.198) and 2 mm (mΔF terylene2 = 8.71°; mΔF steel2 = 10.41°; p = 0.114). Elastic deformation of terylene was significantly greater than that of steel (p = 0.0004).


CONCLUSIONS
Suture protection of the patellar ligament by a terylene wire is not significantly different from that provided by steel frame. The elastic properties of terylene and absence of a need for re intervention to secure its removal lead us towards its use in acute ruptures of the patellar ligament. The main limits involve the properties of the chain extenders with no contraction/muscle shortening and partial dehydration of tendons and ligaments and the mean age of 69.3 years. Level 5.
Department of Orthopedics and Traumatology, University Hospital Center of Poitiers, Poitiers, France. pbouget@hotmail.fr.
P
P
Bouget
ABS Lab, University School of Medicine of Poitiers, Poitiers, France.
C
C
Breque
Department of Orthopedics and Traumatology, André Mignot Hospital, Hospital Center of Versailles, Versailles, France.
J S
JS
Beranger
ABS Lab, University School of Medicine of Poitiers, Poitiers, France.|Department of Visceral Surgery, University Hospital Center of Poitiers, Poitiers, France.
J P
JP
Faure
Department of Orthopedics and Traumatology, La Pitié Salpêtrière Hospital, Public Hospital of Paris, Paris, France.
F
F
Khiami
Department of Orthopedics and Traumatology, University Hospital Center of Poitiers, Poitiers, France.
T
T
Vendeuvre
Frame; Patellar ligament; Patellar tendon; Polyester; Rupture; Terylene