Reference-free spectroscopic determination of fat and protein in milk in the visible and near infrared region below 1000nm using spatially resolved diffuse reflectance fiber probe.
New technique of diffuse reflectance spectroscopic analysis of milk fat and total protein content in the visible (Vis) and adjacent near infrared (NIR) region (400-995nm) has been developed and tested. Sample analysis was performed through a probe having eight 200-µm fiber channels forming a linear array. One of the end fibers was used for the illumination and other seven - for the spectroscopic detection of diffusely reflected light. One of the detection channels was used as a reference to normalize the spectra and to convert them into absorbance-equivalent units. The method has been tested experimentally using a designed sample set prepared from industrial raw milk standards with widely varying fat and protein content. To increase the modelling robustness all milk samples were measured in three different homogenization degrees. Comprehensive data analysis has shown the advantage of combining both spectral and spatial resolution in the same measurement and revealed the most relevant channels and wavelength regions. The modelling accuracy was further improved using joint variable selection and preprocessing optimization method based on the genetic algorithm. The root mean-square errors of different validation methods were below 0.10% for fat and below 0.08% for total protein content. Based on the present experimental data, it was computationally shown that the full-spectrum analysis in this method can be replaced by a sensor measurement at several specific wavelengths, for instance, using light-emitting diodes (LEDs) for illumination. Two optimal sensor configurations have been suggested: with nine LEDs for the analysis of fat and seven - for protein content. Both simulated sensors exhibit nearly the same component determination accuracy as corresponding full-spectrum analysis.
J&M Analytik AG, 8 Willy-Messerschmitt-Straße, 73457 Essingen, Germany; Samara State Technical University, Molodogvardeyskaya Street 244, 443100 Samara, Russia. Electronic address: ab@globalmodelling.com.
Andrey
A
Bogomolov
Samara State Technical University, Molodogvardeyskaya Street 244, 443100 Samara, Russia.
Valeria
V
Belikova
Samara State Technical University, Molodogvardeyskaya Street 244, 443100 Samara, Russia.
Vladislav
V
Galyanin
Samara State Technical University, Molodogvardeyskaya Street 244, 443100 Samara, Russia.
Anastasiia
A
Melenteva
J&M Analytik AG, 8 Willy-Messerschmitt-Straße, 73457 Essingen, Germany.
Hans
H
Meyer
D000818:Animals; D002417:Cattle; D005223:Fats / Q000032:analysis*; D005336:Fiber Optic Technology / Q000379:methods*; D008892:Milk / Q000737:chemistry*; D008894:Milk Proteins / Q000032:analysis*; D010783:Photometry / Q000379:methods*; D012015:Reference Standards; D019265:Spectroscopy, Near-Infrared / Q000379:methods*
D005223:Fats; D008894:Milk Proteins
Milk nutrient analysis; Optical sensor; Space-resolved measurement; Three-way PLS regression; Variable selection; Vis/NIR spectroscopy