title: High False Positives and False Negatives in Yeast Parameter in an Automated Urine Sediment Analyzer.
abstract: BACKGROUND
  Automated urine sediment analyzers have proven their feasibility in medical laboratories. However, editing manual microscopic review of some specimens severely limits the usefulness of such systems. This study aims to give feedback on the practical experience on »Yeast«, which is one of the parameters that compel frequent manual reviews.
  
  
  METHODS
  5448 freshly collected urine specimens submitted from various departments of our hospital for diagnostic urinalysis were studied by the UriSed® (77 Elektronika, Hungary). A specialist medical doctor inspected every image on-board, and reviewed the ones with a »Yeast« alarm by traditional manual microscopy.
  
  
  RESULTS
  UriSed alarmed in 491 samples (9%) for yeast. In 59 samples (1%) the number of particles exceeded the cut-off and a »positive for yeast« was set. A false positive report of yeast +1 to 3+/HPF was found in 51 samples (0.9%). There were 8 cases with positive for yeast from both microscopic methods. Thirty-three »negative for yeast« samples were corrected as positive after the manual microscopic review.
  
  
  CONCLUSIONS
  We report a high percentage of false positives and negatives in the yeast parameter, in line with other studies on UriSed as well as on other instruments in the market. As an important feedback, our observations showed that the major concern in false results was »the focusing problem«. We believe in the necessity of a focus check and comparison of alarms between images on board.
authors.affiliation: Clinical Biochemistry, Batman Maternity and Children's Hospital, Batman, Turkey.
authors.forename: Ozgur
authors.initials: O
authors.lastname: Aydin
authors.affiliation: Central Laboratories, Antalya Education and Research Hospital, Antalya, Turkey.
authors.forename: Hamit Yasar
authors.initials: HY
authors.lastname: Ellidag
authors.affiliation: Clinical Biochemistry, Atatürk Hospital, Antalya, Turkey.
authors.forename: Esin
authors.initials: E
authors.lastname: Eren
authors.affiliation: Central Laboratories, Antalya Education and Research Hospital, Antalya, Turkey.
authors.forename: Necat
authors.initials: N
authors.lastname: Yilmaz
keywords: Urised/SediMAX; automated urine analysis; urine; urine microscopy; yeast